NSF Product and Service Listings
These NSF Official Listings are current as of Monday, December 05, 2016 at 12:15 a.m. Eastern Time. Please contact NSF International to confirm the status of any Listing, report errors, or make suggestions.
Alert: NSF is concerned about fraudulent downloading and manipulation of website text. Always confirm this information by clicking on the below link for the most accurate information: http://info.nsf.org/Certified/Biosafety/Listings.asp?
Class II (Laminar Flow) Biosafety Cabinetry
Cabinet Style A = Bench Unit With Base or Adjustable Legs Provided B = Bench Unit Without Base or Adjustable Legs Provided C = Console 
Window Type H = Hinged S = Sliding F = Fixed 
Bench Height NA = Not Applicable NP = Not Provided 
CBV = Concurrent Balance Value
Due to a change in nomenclature in NSF/ANSI 49  2002 “Class II (Laminar Flow) Biosafety Cabinetry,” Class II, Type A cabinets are now referred to as Class II, Type A1 and Class II, Types B3 and A/B3 cabinets as Class II, Type A2. Class II, Types B1 and B2 cabinets remain unchanged.
Biosafety cabinet models Listed under NSF/ANSI 49 are approved for use with a power supply of 115V/60Hz, unless otherwise noted. If biosafety cabinets have been approved for use with additional power supplies, the power supply will be indicated in the Official Listing.
Allentown, Inc.  
Route 526  
P.O. Box 698  
Allentown, NJ 085010698  
United States  
6092597951  
Visit this company's website 
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

39241[1] [2]  A2  A  100  110  50  60  N/A  4  10S  36  Canopy Connection I.V. Pole U.V. Light 
39242 Safety Cabinet[1] [3]  A2  A  100  110  55  65  N/A  4  12S  36  Canopy Connection I.V. Pole U.V. Light 
39248[1] [2]  A2  A  100  110  50  60  N/A  4  8S  36  Canopy Connection I.V. Pole U.V. Light 
39251[1] [2]  A2  A  100  110  50  60  N/A  5  10S  36  Canopy Connection I.V. Pole U.V. Light 
39252[1] [3]  A2  A  100  110  55  65  N/A  5  12S  36  Canopy Connection I.V. Pole U.V. Light 
39258[1] [2]  A2  A  100  110  50  60  N/A  5  8S  36  Canopy Connection I.V. Pole U.V. Light 
39261[1] [2]  A2  A  100  110  50  60  N/A  6  10S  36  Canopy Connection I.V. Pole U.V. Light 
39262 Safety Cabinet[1] [3]  A2  A  100  110  55  65  N/A  6  12S  36  Canopy Connection I.V. Pole U.V. Light 
39268[1] [2]  A2  A  100  110  50  60  N/A  6  8S  36  Canopy Connection I.V. Pole U.V. Light 
[1] Certified for use with power supplies of 100V/50Hz, 100V/60Hz, 230V/50Hz, and 230V/60Hz. [2] Inflow nominal set point of 105 fpm was established with a direct airflow reading measurement. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted access opening, using the manufacturersupplied probe holder and sash stop (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 55 fpm was established with I.V. pole and U.V. light removed. This cabinet model was Certified to NSF/ANSI 492012. This cabinet model is Certified for use with optional sidewall VacuPassTM Cord and Cable Portal. [3] Inflow nominal set point of 105 fpm was established with a direct airflow reading measurement. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted access opening, using the manufacturersupplied probe holder and sash stop (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 60 fpm was established with I.V. pole and U.V. light removed. This cabinet model was Certified to NSF/ANSI 492014. This cabinet model is Certified for use with optional sidewall VacuPassTM Cord and Cable Portal. 
Baker Company, Inc. (The)  
Sanford Airport  
P.O. Drawer E  
Sanford, ME 04073  
United States  
8009922537  
2073248773  
Visit this company's website 
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

BCG401[1]  B2  C  100  110  35  45  664 @ 1.6  4  8S  35  I.V. Pole U.V. Light Casters 
BCG601[1]  B2  C  100  110  35  45  993 @ 1.6  6  8S  35  I.V. Pole U.V. Light 
NCB405[3]  B1  C  100  110  65  75 50  60 
276 @ 0.7  4  8S  NA  I.V. Pole U.V. Light 
NCB405INT[3] [4]  B1  C  100  110  65  75 50  60 
276 @ 0.7  4  8S  NA  I.V. Pole U.V. Light 
NCB605[3]  B1  C  100  110  65  75 50  60 
443 @ 0.8  6  8S  NA  I.V. Pole U.V. Light 
NCB605INT[3] [4]  B1  C  100  110  65  75 50  60 
443 @ 0.8  6  8S  NA  I.V. Pole U.V. Light 
SG404[2] [4] [5] [6]  A2  A  100  110  47  57 42  52 
N/A  4  8S  37  I.V. Pole U.V. Light Casters Canopy Connection 
SG404[2] [4] [5] [6]  A2  A  100  110  47  57 42  52 
N/A  4  10S  37  I.V. Pole U.V. Light Casters Canopy Connection 
SG404[2] [4] [5] [6]  A2  A  100  110  47  57 42  52 
N/A  4  12S  37  I.V. Pole U.V. Light Casters Canopy Connection 
SG504[2] [4] [6] [7]  A2  A  100  110  45  55 42  52 
N/A  5  8S  37  I.V. Pole U.V. Light Casters Canopy Connection 
SG504[2] [4] [6] [7]  A2  A  100  110  45  55 42  52 
N/A  5  10S  37  I.V. Pole U.V. Light Casters Canopy Connection 
SG504[2] [4] [6] [8]  A2  A  100  110  45  55 40  50 
N/A  5  12S  37  I.V. Pole U.V. Light Casters Canopy Connection 
SG604[2] [4] [6] [9]  A2  A  100  110  48  58 42  52 
N/A  6  8S  37  Casters Canopy Connection I.V. Pole U.V. Light 
SG604[2] [4] [6] [9]  A2  A  100  110  48  58 42  52 
N/A  6  10S  37  Casters Canopy Connection I.V. Pole U.V. Light 
SG604[2] [4] [6] [10]  A2  A  100  110  42  52 36  46 
N/A  6  12S  37  Canopy Connection Casters I.V. Pole U.V. Light 
[1] Beginning with serial number 103539 for the 4foot model and serial number 103706 for the 6foot model. Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate calculated method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer's operator's manual for appropriate correction factor), without adjusting cabinet airflow balance. Downflow nominal set point was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492010a. This cabinet qualifies only with the armrest installed. This cabinet is approved with additional petcocks on the sidewalls and a low static pressure exhaust plenum. Cabinet models BCG401 INT and BCG601 INT are Certified for alternate motor/blower combinations and input power supply. [2] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate calculated method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer's operator's manual for appropriate correction factor, if applicable). [3] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate calculated method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Zoned downflow nominal set points of 70 fpm (zone 1  comprised of the rear and middle rows) and 55 fpm (zone 2  the row closest the window opening) were established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492011. NCBX05 model approved with prefilters located under the work surface placed over the primary supply filters. [4] Baker model designations SG404AUS and SG604AUS are Certified to NSF/ANSI 492012 for operation at 220V/50Hz with an 8inch window opening. Baker model designations SG404INT, SG504INT, and SG604INT are Certified to NSF/ANSI 492012 for operation at either 220V/50Hz or 220V/60Hz. [5] Beginning with serial number 110235. The acceptable nonuniform downflow ranges listed above begin with zone 1 (4757 fpm) at the rear of the work area (compromised of lateral rear and center rows) and end with zone 2 (4252 fpm) (comprised of front lateral row) behind the front window enclosing the work area. Downflow nominal set points were established with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. This cabinet qualifies only with the armrest installed. FlexAir Canopy connection 333A745 is approved for use with this model. [6] The maximum height for the bench stand with casters is 30 inches. The maximum height for the bench stand with feet is 37 inches. [7] Beginning with serial number 110795. The acceptable nonuniform downflow ranges listed above begin with zone 1 (4555 fpm) at the rear of the work area (compromised of lateral rear and center rows) and end with zone 2 (4252 fpm) (comprised of front lateral row) behind the front window enclosing the work area. Downflow nominal set points were established with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. This cabinet qualifies only with the armrest installed. FlexAir Canopy connection 336A745 is approved for use with this model. [8] Beginning with serial number 110795. The acceptable nonuniform downflow ranges listed above begin with zone 1 (4555 fpm) at the rear of the work area (compromised of lateral rear and center rows) and end with zone 2 (4050 fpm) (comprised of front lateral row) behind the front window enclosing the work area. Downflow nominal set points were established with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. This cabinet qualifies only with the armrest installed. FlexAir Canopy connection 336A745 is approved for use with this model. [9] Beginning with serial number 110555. The acceptable nonuniform downflow ranges listed above begin with zone 1 (4858 fpm) at the rear of the work area (compromised of lateral rear and center rows) and end with zone 2 (4252 fpm) (comprised of front lateral row) behind the front window enclosing the work area. Downflow nominal set points were established with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. This cabinet qualifies only with the armrest installed. FlexAir Canopy connection 336A745 is approved for use with this model. [10] Beginning with serial number 110555. The acceptable nonuniform downflow ranges listed above begin with zone 1 (4252 fpm) at the rear of the work area (compromised of lateral rear and center rows) and end with zone 2 (3646 fpm) (comprised of front lateral row) behind the front window enclosing the work area. Downflow nominal set points were established with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. This cabinet qualifies only with the armrest installed. FlexAir Canopy connection 336A745 is approved for use with this model. 
BIOBASE Biodustry (Shandong) Co., Ltd.  
No. 2717 First Industry Road  
Mingshui Economic Development Zone  
Zhangqiu, Jinan, Shandong Province  
China  
86 531 8121 9800  
Visit this company's website 
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

11235 bbc 86[1] [2]  A2  A  100  110  60  70  N/A  3  8S  33  I.V. Pole 
11236 bbc 86[1] [2]  A2  A  100  110  60  70  N/A  6  8S  33  I.V. Pole 
BSC4FA2[2] [3]  A2  A  100  110  60  70  N/A  4  8S  27  I.V. Pole U.V. Light 
[1] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate calculated method with thermal anemometer in a constricted (80 mm high) access opening (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 65 fpm was established with I.V. Pole removed. This cabinet model was Certified to NSF/ANSI 492011. [2] Approved for alternate power modes of 220V/50Hz and 220V/60Hz. [3] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal setpoint was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal setpoint of 65 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. 
CHC Lab  
5201 Youngsandong  
Yusunggu  
Daejeon 305500  
Republic of Korea  
82 42 933 0036  
Visit this company's website 
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

CHC777A204[1] [3]  A2  A  100  110  65  75  N/A  4  8S  33  Casters I.V. Pole U.V. Light 
CHC777A206[1] [4]  A2  A  100  110  65  75  N/A  6  8S  33  Casters I.V. Pole U.V. Light 
CHC888B204[2] [4]  B2  A  100  110  60  70  818 @ 2.4  4  8S  33  I.V. Pole U.V. Light 
CHC888B206[4] [5]  B2  A  100  110  55  65  1231@3.16  6  8S  32  I.V. Pole U.V. Light 
[1] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's alternate recomemded method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer's operator's manual for appropriate correction factor), without adjusting cabinet airflow balance. Downflow nominal set point was established with I.V.Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492007. This cabinet qualifies only with the armrest installed. [2] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's alternate recomemded method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer's operator's manual for appropriate correction factor), without adjusting cabinet airflow balance. Downflow nominal set point was established with I.V.Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492009. [3] Certified for use with power supplies of 115V/60 Hz, 220V/50 Hz, and 220V/60 Hz. [4] Certified for use with power supplies of 115V/60 Hz and 220V/60 Hz. [5] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's alternate recommended method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer's operator's manual for appropriate correction factor), without adjusting cabinet airflow balance. Downflow nominal set point was established with I.V.Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492010. This cabinet qualifies only with the armrest installed. 
Chung Fu Technical Development Co., Ltd.  
7F, No. 687, Mintsu East Road  
Taipei 105  
Taiwan  
886 2 2600 1761  
Visit this company's website 
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

AHA143AAA[1] [2]  A2  A  100  110  68  78  N/A  4  8S  26  I.V. Pole U.V. Light 
AHA143ABA[1] [2] [3]  A2  A  100  110  68  78  N/A  4  8S  26  I.V. Pole U.V. Light 
AHA143BAA[1] [2]  A2  A  100  110  68  78  N/A  4  8S  26  I.V. Pole U.V. Light 
AHA143BBA[1] [2] [3]  A2  A  100  110  68  78  N/A  4  8S  26  I.V. Pole U.V. Light 
[1] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's alternate recomended method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer's operator's manual for appropriate correction factor), without adjusting cabinet airflow balance. Downflow nominal set point was established with I.V.Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492009. [2] Chung Fu Technical Development Co., Ltd. Models AHA143BAA, AHA143BBA, AHA143AAA and AHA143ABA are marketed by Air Science, LLC, USA as Air Science Models PB48R115 (115V, 60Hz), PB48R230 (230V, 50Hz), PB48M115 (115V, 60Hz) and PB48M230 (230V, 50Hz). [3] Certified for use with power supply of 220V/50Hz. 
Design Filtration Microzone Inc.  
86 Harry Douglas Drive  
Ottawa, ON K2S 2C7  
Canada  
6137025017  
Visit this company's website 
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

BK34[1] [2]  A2  A  100  110  60  70  N/A  4  8S  27  I.V. Pole U.V. Light 
[1] Approved for alternate power modes of 220V/50Hz and 220V/60Hz. [2] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal setpoint was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal setpoint of 65 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. 
Esco Micro Pte Ltd  
21 Changi South Street 1  
486777  
Singapore  
65 6542 0833  
Visit this company's website 
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

AC23S8NS[1] [2] [3]  A2  A  100  110  55  65  N/A  3  8S  35  I.V. Pole U.V. Light Canopy Connection 
AC23S9NS[1]  A2  A  100  100  55  65  N/A  3  8S  35  I.V. Pole U.V. Light Canopy Connection 
AC24N7[4] [5]  A2  A  100  110  55  65  N/A  4  8S  36  I.V. Pole U.V. Light 
AC24S8NS[2] [3] [4]  A2  A  100  110  55  65  N/A  4  8S  36  I.V. Pole U.V. Light 
AC24S9NS[4]  A2  A  100  110  55  65  N/A  4  8S  36  I.V. Pole U.V. Light 
AC24Y7[4] [5]  A2  A  100  110  55  65  N/A  4  8S  36  I.V. Pole U.V. Light 
AC25S8NS[2] [3] [6]  A2  A  100  110  55  65  N/A  5  8S  35  I.V. Pole U.V. Light 
AC25S9NS[6]  A2  A  100  110  55  65  N/A  5  8S  35  I.V. Pole U.V. Light 
AC26N7[5] [7]  A2  A  100  110  55  65  N/A  6  8S  36  I.V. Pole U.V. Light 
AC26S8NS[2] [3] [7]  A2  A  100  110  55  65  N/A  6  8S  36  I.V. Pole U.V. Light 
AC26S9NS[7]  A2  A  100  110  55  65  N/A  6  8S  36  I.V. Pole U.V. Light 
AC26Y7[5] [7]  A2  A  100  110  55  65  N/A  6  8S  36  I.V. Pole U.V. Light 
AR23S8[1] [2] [3]  A2  A  100  110  55  65  N/A  3  8S  35  I.V. Pole U.V. Light Canopy Connection 
AR23S9[1]  A2  A  100  110  55  65  N/A  3  8S  35  I.V. Pole U.V. Light Canopy Connection 
AR24S9[4]  A2  A  100  110  55  65  N/A  4  8S  36  I.V. Pole U.V. Light 
AR25S8[2] [3] [6]  A2  A  100  110  55  65  N/A  5  8S  35  I.V. Pole U.V. Light 
AR25S9[6]  A2  A  100  110  55  65  N/A  5  8S  35  I.V. Pole U.V. Light 
AR26S9[7]  A2  A  100  110  55  65  N/A  6  8S  36  I.V. Pole U.V. Light 
BC24S7[4] [5]  A2  A  100  100  55  65  N/A  4  8S  36  I.V. Pole U.V. Light 
BC26S7[5] [7]  A2  A  100  110  55  65  N/A  6  8S  36  I.V. Pole U.V. Light 
LA24A1E[2] [8]  A2  A  100  110  65  75  N/A  4  9S  34  I.V. Pole U.V. Light 
LA24A2E[8]  A2  A  100  110  65  75  N/A  4  9S  34  I.V. Pole U.V. Light 
LA24A3E[3] [8]  A2  A  100  110  65  75  N/A  4  9S  34  I.V. Pole U.V. Light 
LA25A1E[2] [8]  A2  A  100  110  65  75  N/A  5  9S  36  I.V. Pole U.V. Light 
LA25A2E[8]  A2  A  100  110  65  75  N/A  5  9S  36  I.V. Pole U.V. Light 
LA25A3E[3] [8]  A2  A  100  110  65  75  N/A  5  9S  36  I.V. Pole U.V. Light 
LA26A1E[2] [8]  A2  A  100  110  60  70  N/A  6  8S  35  I.V. Pole U.V. Light 
LA26A2E[8]  A2  A  100  110  60  70  N/A  6  8S  35  I.V. Pole U.V. Light 
LA26A3E[3] [8]  A2  A  100  110  60  70  N/A  6  8S  35  I.V. Pole U.V. Light 
LB24B1E[2] [9]  B2  A  100  110  55  65  803 @ 1.9  4  8S  36  I.V. Pole U.V. Light 
LB24B2E[9]  B2  A  100  110  55  65  803 @ 1.9  4  8S  36  I.V. Pole U.V. Light 
LB24B3E[3] [9]  B2  A  100  110  55  65  803 @ 1.9  4  8S  36  I.V. Pole U.V. Light 
LB25B1E[2] [9] [10]  B2  A  100  110  55  65  1025 @ 1.8  5  8S  35  I.V. Pole U.V. Light 
LB25B2E[9] [10]  B2  A  100  110  55  65  1025 @ 1.8  5  8S  35  I.V. Pole U.V. Light 
LB25B3E[3] [9] [10]  B2  A  100  110  55  65  1025 @ 1.8  5  8S  35  I.V. Pole U.V. Light 
LB26B1E[2] [9]  B2  A  100  110  55  65  1269 @ 2.1  6  8S  36  I.V. Pole U.V. Light 
LB26B2E[9]  B2  A  100  110  55  65  1269 @ 2.1  6  8S  36  I.V. Pole U.V. Light 
LB26B3E[3] [9]  B2  A  100  110  55  65  1269 @ 2.1  6  8S  36  I.V. Pole U.V. Light 
LR24S1E[2] [8]  A2  A  100  110  65  75  N/A  4  9S  34  I.V. Pole U.V. Light 
LR24S2E[8]  A2  A  100  110  65  75  N/A  4  9S  34  I.V. Pole U.V. Light 
LR24S3E[3] [8]  A2  A  100  110  65  75  N/A  4  9S  34  I.V. Pole U.V. Light 
LR25S1E[2] [8]  A2  A  100  110  65  75  N/A  5  9S  36  I.V. Pole U.V. Light 
LR25S2E[8]  A2  A  100  110  65  75  N/A  5  9S  36  I.V. Pole U.V. Light 
LR25S3E[3] [8]  A2  A  100  110  65  75  N/A  5  9S  36  I.V. Pole U.V. Light 
LR26S1E[2] [8]  A2  A  100  110  60  70  N/A  6  8S  35  I.V. Pole U.V. Light 
LR26S2E[8]  A2  A  100  110  60  70  N/A  6  8S  35  I.V. Pole U.V. Light 
LR26S3E[3] [8]  A2  A  100  110  60  70  N/A  6  8S  35  I.V. Pole U.V. Light 
[1] Beginning with serial number 2015  94341. Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 60 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492014. [2] Certified for use with a power supply of 230V/50Hz. [3] Certified for use with a power supply of 230V/60Hz. [4] Beginning with serial number 2014  89128. Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 60 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. [5] Approved for alternate power modes of 100V/50Hz and 100V/60Hz. [6] Beginning with serial number 2015  94651. Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 60 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492014. [7] Beginning with serial number 2014  84641. Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 60 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. [8] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 70 fpm was established with I.V. Pole and U.V. Light removed.This cabinet model was Certified to NSF/ANSI 492011. [9] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 60 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. [10] Certified for use as benchtop model or with SPL5B0 stand only. 
Kewaunee Scientific Corporation  
2700 West Front Street  
Statesville, NC 28677  
United States  
7048737202  
Visit this company's website 
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

INT1100[1] [2]  A2  A  100  110  60  70  N/A  3  10S  33  I.V. Pole U.V. Light 
INT14002[3] [6]  A2  A  100  110  50  60  N/A  4  10S  35  I.V. Pole U.V. Light 
INT1700[2] [4]  A2  A  100  110  55  65  N/A  5  10S  33  I.V. Pole U.V. Light 
INT20002[3] [6]  A2  A  100  110  50  60  N/A  6  10S  35  I.V. Pole U.V. Light 
INTB1400[2] [5]  B2  A  100  110  55  65  967 @ 1.7  4  8S  27  I.V. Pole U.V. Light 
INTB2000[2] [5]  B2  A  100  110  55  65  1256 @ 1.9  6  8S  34  I.V. Pole U.V. Light 
[1] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. This nominal setpoint was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer for appropriate correction factor, if applicable) without adjusting cabinet airflow balance. Downflow nominal setpoint of 65 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492009. [2] Certified for use with power supplies of 115V/60Hz and 230V/50Hz. Models may be followed by 230 suffix. [3] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. This nominal setpoint was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer for appropriate correction factor, if applicable) without adjusting cabinet airflow balance. Downflow nominal setpoint of 55 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model qualifies only with the armrest installed. This cabinet model was Certified to NSF/ANSI 492014. [4] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. This nominal setpoint was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer for appropriate correction factor, if applicable) without adjusting cabinet airflow balance. Downflow nominal setpoint of 60 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model qualifies only with the armrest installed. This cabinet model was Certified to NSF/ANSI 492009. [5] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. This nominal setpoint was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer for appropriate correction factor, if applicable) without adjusting cabinet airflow balance. Downflow nominal setpoint of 60 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model qualifies only with the armrest installed. This cabinet model was Certified to NSF/ANSI 492010a. [6] Also Certified for use with power supplies of 230V/50Hz and 230V/60Hz. 
Labconco Corporation  
8845 Prospect  
Kansas City, MO 64132  
United States  
8163338811 X771  
Visit this company's website 
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

30231 Logic+ Series[1] [2]  A2  A  100  110  50  60  N/A  3  10S  36  I.V. Pole U.V. Light Canopy Connection 
30238 Logic+ Series[1] [3]  A2  A  100  110  50  60  N/A  3  8S  36  I.V. Pole U.V. Light Canopy Connection 
30241 Logic+ Series[3] [4]  A2  A  100  110  50  60  N/A  4  10S  36  Canopy Connection I.V. Pole U.V. Light 
30242 Logic+ Series[5] [6]  A2  A  100  110  55  65  N/A  4  12S  36  Canopy Connection I.V. Pole U.V. Light 
30248 Logic+ Series[3] [4]  A2  A  100  110  50  60  N/A  4  8S  36  Canopy Connection I.V. Pole U.V. Light 
30248xx80 Logic+Series[7]  A2  A  100  110  65  75  N/A  4  8S  36  I.V. Pole U.V. Light 
30251 Logic+ Series[3] [8]  A2  A  100  110  50  60  N/A  5  10S  36  I.V. Pole U.V. Light Canopy Connection 
30252 Logic+ Series[3] [9]  A2  A  100  110  55  65  N/A  5  12S  36  Canopy Connection I.V. Pole U.V. Light 
30258 Logic+ Series[3] [8]  A2  A  100  110  50  60  N/A  5  8S  36  Canopy Connection I.V. Pole U.V. Light 
30261 Logic+ Series[3] [10]  A2  A  100  110  50  60  N/A  6  10S  36  Canopy Connection I.V. Pole U.V. Light 
30262 Logic+ Series[5] [11]  A2  A  100  110  55  65  N/A  6  12S  36  Canopy Connection I.V. Pole U.V. Light 
30268 Logic+ Series[3] [10]  A2  A  100  110  50  60  N/A  6  8S  36  Canopy Connection I.V. Pole U.V. Light 
30348 Logic+ Series[3] [12]  B2  A  100  110  50  60  734 @ 1.7  4  8S  36  BagIn/ BagOut I.V. Pole U.V. Light 
30368 Logic+ Series[3] [13]  B2  A  100  110  50  60  1191 @ 2.1  6  8S  36  BagIn/ BagOut I.V. Pole U.V. Light 
30448 Axiom Series[3] [14] [15]  NA  A  100  110  60  70  N/A  4  8S  36  I.V. Pole U.V. Light BagIn/ BagOut 
30441 Axiom Series[3] [14] [15]  NA  A  100  110  60  70  N/A  4  10S  36  I.V. Pole U.V. Light BagIn/ BagOut 
30468 Axiom Series[3] [15] [16]  NA  A  100  110  50  60  N/A  6  8S  36  I.V. Pole U.V. Light BagIn/ BagOut 
30461 Axiom Series[3] [15] [16]  NA  A  100  110  50  60  N/A  6  10S  36  I.V. Pole U.V. Light BagIn/ BagOut 
31261 Logic+ PuriCare[3] [10]  A2  A  100  110  50  60  N/A  6  10S  36  I.V. Pole U.V. Light Canopy Connection 
[1] Beginning with serial number 130300000. Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted access opening, using the manufacturersupplied probe holder and sash stop (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 55 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492011. This cabinet model is Certified for use with optional sidewall VacuPassTM Cord & Cable Portal. [2] Certified for use with power supplies of 230V/50 Hz and 230V/60 Hz. [3] Certified for use with power supplies of 100V/50 Hz, 100V/60 Hz, 230V/50 Hz, and 230V/60 Hz. [4] Beginning with serial number 130300000. Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted access opening, using the manufacturersupplied probe holder and sash stop (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 55 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. This cabinet model is Certified for use with optional sidewall VacuPassTM Cord & Cable Portal. [5] Certified for use with power supplies of 100V/50 Hz, 100V/60 Hz, and 230V/50 Hz. [6] Beginning with serial number 150800001. Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted access opening, using the manufacturersupplied probe holder and sash stop (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 60 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492014. This cabinet model is Certified for use with optional sidewall VacuPassTM Cord & Cable Portal. [7] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. Inflow nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted access opening, using the manufacturersupplied probe holder and sash stop (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 70 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. This cabinet model is Certified for use with optional sidewall VacuPassTM Cord & Cable Portal. Certified for use with power supply of 220V/50Hz only. Model designations of 302480080, 302480180, 302481080, and 302481180 are all aceptable, representing different configurations of NSF acceptable options. [8] Beginning with serial number 130300000. Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted access opening, using the manufacturersupplied probe holder and sash stop (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 55 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. This cabinet model is Certified for use with optional sidewall VacuPassTM Cord & Cable Portal. [9] Beginning with serial number 151200002. Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted access opening, using the manufacturersupplied probe holder and sash stop (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 60 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492014. This cabinet model is Certified for use with optional sidewall VacuPassTM Cord & Cable Portal. [10] Beginning with serial number 130300000. Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted access opening, using the manufacturersupplied probe holder and sash stop (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 55 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. This cabinet model is Certified for use with optional sidewall VacuPassTM Cord & Cable Portal. [11] Beginning with serial number 15100000. Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted access opening, using the manufacturersupplied probe holder and sash stop (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 60 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492014. This cabinet model is Certified for use with optional sidewall VacuPassTM Cord & Cable Portal. [12] Beginning with serial number 130500000. Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted access opening, using the manufacturersupplied probe holder and sash stop (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 55 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. This cabinet model is Certified for use with optional sidewall VacuPassTM Cord & Cable Portal. [13] Beginning with serial number 130500000. Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted access opening, using the manufacturersupplied probe holder and sash stop (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 55 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. This cabinet model is Certified for use with optional sidewall VacuPassTM Cord & Cable Portal. [14] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. Inflow nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted access opening, using the manufacturersupplied probe holder and sash stop (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 65 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. [15] The design of this class II biosafety cabinet does not fall under the current definitions in NSF/ANSI 492012 for type A or type B cabinets. However, this biosafety cabinet meets the materials, design and construction, and performance requirements of NSF/ANSI 492012 and is Certified by NSF International. [16] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. Inflow nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted access opening, using the manufacturersupplied probe holder and sash stop (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 55 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. 
NOTE: Bench stand models 3730300, 3730310, 3780200 and 3780100 may be used with cabinets in the 30231/30238 Logic+ series. Bench stand models 3730400, 3730410, 3780201 and 3780101 may be used with cabinets in the 30241/30248/30242 Logic+ series and the 31242 PuriCare series. Bench stand models 3730500, 3730510, 3780203, 3780106, 3780107, 3780312, and 3780316 may be used with cabinets in the 30251/30258 Logic+ series. Bench stand models 3730600, 3630610, 3780202 and 3780102 may be used with cabinets in the 30261/30268/30262 Logic+ series and the 31262 PuriCare series. Bench stand model 3730400 may also be used with the 30348 Logic+ Series. All bench stand models beginning with 378 may be used with or without the optional wheel kit (PN# 3784000).
MEDFUTURE LLC  
97 East Brokaw Road  
Suite 240  
San Jose, CA 95112  
United States  
6262834740  
Visit this company's website 
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

BIIA23F[1] [2]  A2  A  100  110  60  70  N/A  3  8S  33  I.V. Pole 
BIIA24F[2] [3]  A2  A  100  110  60  70  N/A  4  8S  27  I.V. Pole U.V. Light 
BIIA26F[1] [2]  A2  A  100  110  60  70  N/A  6  8S  33  I.V. Pole 
[1] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate calculated method with thermal anemometer in a constricted (80 mm high) access opening (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 65 fpm was established with I.V. Pole removed. This cabinet model was Certified to NSF/ANSI 492011. [2] Approved for alternate power modes of 220V/50Hz and 220V/60Hz. [3] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal setpoint was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal setpoint of 65 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. 
Nuaire, Inc.  
2100 Fernbrook Lane  
Plymouth, MN 55447  
United States  
7635531270  
Visit this company's website 
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

NU425400[1] [2] [3]  A2  A  100  110  55  65  N/A  4  8S  32  I.V. Pole U.V. Light Canopy Connection 
NU425400[1] [2] [3]  A2  A  100  110  55  65  N/A  4  10S  32  I.V. Pole U.V. Light Canopy Connection 
NU425500[1] [3] [4]  A2  A  100  110  55  65  N/A  5  8S  32  U.V. Light I.V. Pole Canopy Connection 
NU425500[1] [3] [4]  A2  A  100  110  55  65  N/A  5  10S  32  U.V. Light I.V. Pole Canopy Connection 
NU425600[1] [3] [5]  A2  A  100  110  55  65  N/A  6  8S  32  I.V. Pole U.V. Light Canopy Connection 
NU425600[1] [3] [5]  A2  A  100  110  55  65  N/A  6  10S  32  I.V. Pole U.V. Light Canopy Connection 
NU427400[1] [3] [6]  B1  A  100  110  65  75  282 @ 0.7  4  8S  32  BagIn/ BagOut I.V. Pole U.V. Light 
NU427600[1] [3] [7]  B1  A  100  110  65  75  474 @ 0.9  6  8S  32  I.V. Pole U.V. Light BagIn/ BagOut 
NU430400[1] [3] [8] [9]  B2  A  100  110  55  65  785 @ 1.7  4  8S  32  BagIn/ BagOut I.V. Pole U.V. Light 
NU430600[1] [3] [8] [10]  B2  A  100  110  55  65  1250 @ 1.8  6  8S  32  BagIn/ BagOut I.V. Pole U.V. Light 
NU540300[11]  A2  A  100  110  55  65  N/A  3  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU540300[11]  A2  A  100  110  55  65  N/A  3  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU540400[11]  A2  A  100  110  55  65  N/A  4  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU540400[11]  A2  A  100  110  55  65  N/A  4  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU540500[11]  A2  A  100  110  57  67  N/A  5  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU540500[11]  A2  A  100  110  57  67  N/A  5  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU540600[11]  A2  A  100  110  55  65  N/A  6  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU540600[11]  A2  A  100  110  55  65  N/A  6  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU543300[11]  A2  A  100  110  55  65  N/A  3  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU543300[11]  A2  A  100  110  55  65  N/A  3  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU543300[11]  A2  A  100  110  55  65  N/A  3  12S  35  I.V. Pole U.V. Light Canopy Connection 
NU543400[11]  A2  A  100  110  55  65  N/A  4  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU543400[11]  A2  A  100  110  55  65  N/A  4  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU543400[11]  A2  A  100  110  55  65  N/A  4  12S  35  I.V. Pole U.V. Light Canopy Connection 
NU543500[11]  A2  A  100  110  57  67  N/A  5  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU543500[11]  A2  A  100  110  57  67  N/A  5  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU543500[11]  A2  A  100  110  57  67  N/A  5  12S  35  I.V. Pole U.V. Light Canopy Connection 
NU543600[11]  A2  A  100  110  55  65  N/A  6  8S  35  U.V. Light I.V. Pole Canopy Connection 
NU543600[11]  A2  A  100  110  55  65  N/A  6  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU543600[11]  A2  A  100  110  55  65  N/A  6  12S  35  I.V. Pole U.V. Light Canopy Connection 
NU545300[12]  A2  A  100  110  55  65  N/A  3  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU545300[12]  A2  A  100  110  55  35  N/A  3  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU545400[13]  A2  A  100  110  55  65  N/A  3  8S  S  I.V. Pole U.V. Light Canopy Connection 
NU545400[13]  A2  A  100  110  55  65  N/A  4  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU545500[14]  A2  A  100  110  55  65  N/A  5  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU545500[14]  A2  A  100  110  55  65  N/A  5  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU545600[15]  A2  A  100  110  55  65  N/A  6  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU545600[15]  A2  A  100  110  55  65  N/A  6  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU677400[16]  A2  A  100  110  55  65  N/A  4  12S  38  Canopy Connection U.V. Light 
NU677500[17]  A2  A  100  110  55  65  N/A  5  12S  38  Canopy Connection U.V. Light 
NU677600[18]  A2  A  100  110  55  65  N/A  6  12S  38  Canopy Connection U.V. Light 
NUFH677500[19]  A2  A  100  110  55  65  N/A  5  12S  38  U.V. Light 
[1] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening for appropriate correction factor, if applicable, without adjusting cabinet airflow balance. [2] Beginning with series 60 and higher. Downflow nominal setpoint of 60 fpm was established 4 inches above the plane of the window with the I.V. Pole and U.V. Light removed. This cabinet qualifies only with the armrest installed. This cabinet qualifies with optional DC motor installed. This cabinet model was Certified to NSF/ANSI 49â€“2004. Listed model designation NU 437 with electronic control panel and airflow monitor. Listed model designation NU 440 with electronic airflow control system. Listed model may be prefaced with Q, N, L or S (i.e. NUQ425400). [3] Listed model may be suffixed with E. Certified for use with power supplies of 230V/50Hz and 230V/60Hz. [4] Beginning with Series 60 and higher. Downflow nominal setpoint of 60 fpm was established 4 inches above the plane of the window with the I.V. Pole and U.V. Light removed. This cabinet qualifies only with the armrest installed. This cabinet qualifies with optional DC motor installed. This cabinet model was Certified to NSF/ANSI 492004. Listed model designation NU 437 with electronic control panel and airflow monitor. Listed model designation NU 440 with electronic airflow control system. Listed model may be prefaced with Q, N, L, or S (i.e. NUQ425500). [5] Beginning with series 60 and higher. Downflow nominal setpoint of 60 fpm was established 4 inches above the plane of the window with the I.V. Pole and U.V. Light removed. This cabinet qualifies only with the armrest installed. This cabinet qualifies with optional DC motor installed. This cabinet model was Certified to NSF/ANSI 49â€“2004. Listed model designation NU 437 with electronic control panel and airflow monitor. Listed model designation NU 440 with electronic airflow control system. Listed model may be prefaced with Q, N, L or S (i.e. NUQ425600). [6] Beginning with Series 60 and higher. Downflow nominal setpoint of 70 fpm was established 4 inches above the plane of the window with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492011. This cabinet qualifies with the optional DC motor installed. Listed model may be prefaced with Q, N, L or S (i.e. NUQ427400). [7] Beginning with series 40 and higher. Downflow nominal setpoint of 60 fpm was established 4 inches above the plane of the window with the I.V. Pole and U.V. Light removed. This cabinet qualifies with the optional DC motor installed. Listed model may be prefaced with Q, N, L or S (i.e. NUQ427600). This cabinet model was Certified to NSF/ANSI 492011. [8] Listed model designation NU 435 without U.V. Light option. Listed model may be prefaced with Q, N, L or S (i.e. NUQ435400). [9] Beginning with series 60 and higher. The downflow nominal set point of 60 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model qualifies only with the armrest installed. This cabinet qualifies with the optional DC motor installed. This cabinet model was Certified to NSF/ANSI 492009. Listed model may be prefaced with Q, N, L, or S (i.e. NUQ430400). [10] Beginning with series 60 and higher. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and the U.V. Light removed. This cabinet model qualifies only with the armrest installed. This cabinet qualifies with the optional DC motor installed. This cabinet model was Certified to NSF/ANSI 492009. Listed model may be prefaced with Q, N, L, or S (i.e. NUQ430600). [11] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. Listed model may be prefaced with Q, N, L, or S (e.g. NUQ540300). Certified for use with power supplies of 100V/50Hz, 100V/60Hz, 230V/50Hz, and 230V/60Hz. [12] Beginning with serial number 170844102115. Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492014. Listed model may be prefaced with Q, N, L, or S (e.g. NUQ540300). Certified for use with power supplies of 100V/50Hz, 100V/60Hz, 230V/50Hz, and 230V/60Hz. [13] Beginning with serial number 170847102115. Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492014. Listed model may be prefaced with Q, N, L, or S (e.g. NUQ540300). Certified for use with power supplies of 100V/50Hz, 100V/60Hz, 230V/50Hz, and 230V/60Hz. [14] Beginning with serial number 170845102115. Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492014. Listed model may be prefaced with Q, N, L, or S (e.g. NUQ540300). Certified for use with power supplies of 100V/50Hz, 100V/60Hz, 230V/50Hz, and 230V/60Hz. [15] Beginning with serial number 170846102115. Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492014. Listed model may be prefaced with Q, N, L, or S (e.g. NUQ540300). Certified for use with power supplies of 100V/50Hz, 100V/60Hz, 230V/50Hz, and 230V/60Hz. [16] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492011. NuAire canopy model NU911402 was evaluated as part of the NSF Certification testing procedures. [17] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492011. NuAire canopy model NU911502 was evaluated as a part of the NSF Certification testing procedures. [18] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492011. NuAire canopy model NU911602 was evaluated as part of the NSF Certification testing procedures. [19] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. NuAire canopy model NU911502 was evaluated as a part of the NSF Certification testing procedures. NUFH677500 is a modified version of NU677500, including feed hoppers. 
Panasonic Healthcare Corporation of North America  
1300 Michael Drive  
Suite A  
Wood Dale, IL 60191  
United States  
8008588442  
6306948200  
Visit this company's website 
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

MHEN300A2  A2  A  100  110  55  65  N/A  3  10S  35  Canopy Connection I.V. Pole U.V. Light 
MHEN400A2  A2  A  100  110  55  65  N/A  3  10S  35  Canopy Connection I.V. Pole U.V. Light 
MHEN500A2  A2  A  100  110  57  67  N/A  5  10S  35  Canopy Connection I.V. Pole U.V. Light 
MHEN600A2  A2  A  100  110  55  65  N/A  6  10S  35  Canopy Connection I.V. Pole U.V. Light 
[1] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. Listed model may be suffixed with PA, PE, or PP (e.g. MHEN300A2PE). Certified for use with power supplies of 100V/50Hz, 100V/60Hz, 230V/50Hz, and 230V/60Hz.
Shanghai Lishen Scientific Equipment Co., Ltd.  
No. 6788 SongZe Avenue  
QingPu, Shanghai  
China  
86 21 5986 7888  
Visit this company's website 
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

HFsafe1200LC A2[1]  A2  A  100  110  64  74  N/A  4  8S  27  U.V. Light 
HFsafe1200LC N[1] [2]  A2  A  100  110  64  74  N/A  4  8S  27  U.V. Light 
[1] Beginning with serial number 051312LCJ0003EUS. Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (80 mm high) access opening (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow Certified to NSF/ANSI 492012. Approved metric value for cabinet width is 1250 mm, window height is 200 mm, and bench height max is 690 mm. [2] Approved for alternate power mode of 230V/50Hz. 
Suzhou Antai Airtech Co., Ltd.  
No. 2 Weixin Road, SIP  
Suzhou, Jiangsu 215122  
China  
86 512 6552 5002  
Visit this company's website 
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

BSC1301 II A2[1]  A2  A  100  110  64  74  N/A  4  10S  24  U.V. Light Canopy Connection 
BSC1302 II A2[1] [2]  A2  A  100  110  64  74  N/A  4  10S  24  U.V. Light Canopy Connection 
[1] Beginning with serial number J14121740. Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (80 mm high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 69 fpm was established 4 inches above the bottom of the sash with the U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492014. Approved metric value for cabinet width is 1270 mm, window height is 250 mm, and bench height max is 610 mm. [2] Certified for use with power supplies of 230V/50Hz and 230V/60Hz. 
Thermo Fisher Scientific  
RobertBoschStr. 1  
D63505 Langenselbold  
Germany  
49 61 813 5277  
Visit this company's website 
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

1305[1]  A2  A  100  110  58  68  N/A  4  8S  36  Armrest I.V. Pole U.V. Light 
1306[1] [2]  A2  A  100  110  58  58  N/A  4  8S  36  Armrest I.V. Pole U.V. Light 
1307[1]  A2  A  100  110  58  68  N/A  6  8S  36  Armrest I.V. Pole U.V. Light 
1308[1] [2]  A2  A  100  110  58  68  N/A  6  8S  36  Armrest I.V. Pole U.V. Light 
1320[3] [4]  A2  B  100  110  58  68  N/A  3  10S  36  Armrest I.V. Pole U.V. Light 
1321[3] [4]  A2  A  100  110  58  68  N/A  3  10S  36  Armrest I.V. Pole U.V. Light 
1322[3] [4]  A2  B  100  110  58  68  N/A  3  10S  36  Armrest I.V. Pole U.V. Light 
1323[3] [4]  A2  A  100  110  58  68  N/A  3  10S  36  Armrest U.V. Light I.V. Pole 
1325[1]  A2  A  100  110  58  68  N/A  4  8S  36  Armrest I.V. Pole U.V. Light 
1326[1] [2]  A2  A  100  110  58  68  N/A  4  8S  36  Armrest I.V. Pole U.V. Light 
1327[1]  A2  A  100  110  58  68  N/A  6  8S  36  Armrest I.V. Pole U.V. Light 
1328[1] [2]  A2  A  100  110  58  68  N/A  6  8S  36  Armrest I.V. Pole U.V. Light 
1330[2] [3] [4]  A2  B  100  110  58  68  N/A  3  10S  36  Armrest I.V. Pole U.V. Light 
1331[2] [3] [4]  A2  A  100  110  58  68  N/A  3  10S  36  Armrest I.V. Pole U.V. Light 
1332[2] [3] [4]  A2  B  100  110  58  68  N/A  3  10S  36  Armrest I.V. Pole U.V. Light 
1333[2] [3] [4]  A2  A  100  110  58  68  N/A  3  10S  36  Armrest I.V. Pole U.V. Light 
1335[1]  A2  A  100  110  58  68  N/A  4  8S  36  Armrest I.V. Pole U.V. Light 
1336[1] [2]  A2  A  100  110  58  68  N/A  4  8S  36  Armrest I.V. Pole U.V. Light 
1337[1]  A2  A  100  110  58  68  N/A  6  8S  36  Armrest I.V. Pole U.V. Light 
1338[1] [2]  A2  A  100  110  58  68  N/A  6  8S  36  Armrest I.V. Pole U.V. Light 
1339[3] [4]  A2  B  100  110  58  68  N/A  3  8S  36  Armrest I.V. Pole U.V. Light 
1340[3] [4]  A2  A  100  110  58  68  N/A  3  8S  36  Armrest I.V. Pole U.V. Light 
1341[3] [4]  A2  B  100  110  58  68  N/A  3  8S  36  Armrest I.V. Pole U.V. Light 
1342[3] [4]  A2  A  100  110  58  68  N/A  3  8S  36  Armrest I.V. Pole U.V. Light 
1345[5]  A2  A  100  110  58  68  N/A  4  10S  36  Armrest I.V. Pole U.V. Light 
1346[5] [6]  A2  A  100  110  58  68  N/A  4  10S  36  Armrest I.V. Pole U.V. Light 
1347[5]  A2  A  100  110  58  68  N/A  6  10S  36  Armrest I.V. Pole U.V. Light 
1348[5] [6]  A2  A  100  110  58  68  N/A  6  10S  36  Armrest I.V. Pole U.V. Light 
1350[3] [4]  A2  B  100  110  58  68  N/A  5  8S  36  Armrest I.V. Pole U.V. Light 
1351[3] [4]  A2  A  100  110  58  68  N/A  5  8S  36  Armrest I.V. Pole U.V. Light 
1352[3] [4]  A2  B  100  110  58  68  N/A  5  8S  36  Armrest I.V. Pole U.V. Light 
1353[3] [4]  A2  A  100  110  58  68  N/A  5  8S  36  Armrest I.V. Pole U.V. Light 
1354[7]  A2  A  100  110  58  68  N/A  4  10S  36  Armrest I.V. Pole U.V. Light 
1355[5]  A2  A  100  110  58  68  N/A  4  10S  36  Armrest I.V. Pole U.V. Light 
1356[5] [6]  A2  A  100  110  58  68  N/A  4  10S  36  Armrest I.V. Pole U.V. Light 
1357[5]  A2  A  100  110  58  68  N/A  6  10S  36  Armrest I.V. Pole U.V. Light 
1358[5] [6]  A2  A  100  110  58  68  N/A  6  10S  36  Armrest I.V. Pole U.V. Light 
1359[5]  A2  A  100  110  58  68  N/A  6  10S  36  Armrest I.V. Pole U.V. Light 
1366[2] [3] [4]  A2  B  100  110  58  68  N/A  5  8S  36  Armrest I.V. Pole U.V. Light 
1367[2] [3] [4]  A2  A  100  110  58  68  N/A  5  8S  36  Armrest I.V. Pole U.V. Light 
1368[3] [4]  A2  B  100  110  58  68  N/A  5  10S  36  Armrest I.V. Pole U.V. Light 
1369[3] [4]  A2  A  100  110  58  68  N/A  5  10S  36  Armrest U.V. Light I.V. Pole 
1370[3] [4]  A2  B  100  110  58  68  N/A  5  10S  36  Armrest I.V. Pole U.V. Light 
1371[3] [4]  A2  A  100  110  58  68  N/A  5  10S  36  Armrest I.V. Pole U.V. Light 
1372[2] [3] [4]  A2  B  100  110  58  68  N/A  5  8S  36  Armrest U.V. Light I.V. Pole 
1373[2] [3] [4]  A2  A  100  110  58  68  N/A  5  8S  36  Armrest I.V. Pole U.V. Light 
1375[5]  A2  A  100  110  58  68  N/A  4  10S  36  Armrest I.V. Pole U.V. Light 
1376[5] [6]  A2  A  100  110  58  68  N/A  4  10S  36  Armrest I.V. Pole U.V. Light 
1377[5]  A2  A  100  110  58  68  N/A  6  10S  36  Armrest I.V. Pole U.V. Light 
1378[5] [6]  A2  A  100  110  58  68  N/A  6  10S  36  Armrest I.V. Pole U.V. Light 
1380[2] [3] [4]  A2  B  100  110  58  68  N/A  5  10S  36  Armrest I.V. Pole U.V. Light 
1381[2] [3] [4]  A2  A  100  110  58  68  N/A  5  10S  36  Armrest I.V. Pole U.V. Light 
1382[2] [3] [4]  A2  B  100  110  58  68  N/A  5  10S  36  Armrest I.V. Pole U.V. Light 
1383[2] [3] [4]  A2  A  100  110  58  68  N/A  5  10S  36  Armrest I.V. Pole U.V. Light 
1384[7]  A2  A  100  110  58  68  N/A  4  10S  36  Armrest I.V. Pole U.V. Light 
1385[5]  A2  A  100  110  58  68  N/A  4  10S  36  Armrest I.V. Pole U.V. Light 
1386[5] [6]  A2  A  100  110  58  68  N/A  4  10S  36  Armrest I.V. Pole U.V. Light 
1387[5]  A2  A  100  110  58  68  N/A  6  10S  36  Armrest I.V. Pole U.V. Light 
1388[5] [6]  A2  A  100  110  58  68  N/A  6  10S  36  Armrest I.V. Pole U.V. Light 
1389[7]  A2  A  100  110  58  68  N/A  6  10S  36  Armrest I.V. Pole U.V. Light 
1390[2] [3] [4]  A2  B  100  110  58  68  N/A  3  8S  36  Armrest I.V. Pole U.V. Light 
1391[2] [3] [4]  A2  A  100  110  58  68  N/A  3  8S  36  Armrest I.V. Pole U.V. Light 
1392[2] [3] [4]  A2  B  100  110  58  68  N/A  3  8S  36  Armrest I.V. Pole U.V. Light 
1393[2] [3] [4]  A2  A  100  110  58  68  N/A  3  8S  36  Armrest I.V. Pole U.V. Light 
1395[6] [8]  A2  A  100  110  58  68  N/A  4  8S  36  Armrest I.V. Pole U.V. Light 
1396[1] [2]  A2  A  100  110  58  68  N/A  4  8S  36  Armrest I.V. Pole U.V. Light 
1397[6] [8]  A2  A  100  110  58  68  N/A  6  8S  36  Armrest I.V. Pole U.V. Light 
1398[1] [2]  A2  A  100  110  58  68  N/A  6  8S  36  Armrest I.V. Pole U.V. Light 
[1] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal setpoint was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal setpoint of 63 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492009. [2] Certified for use with power supply of 230V/50Hz and 230V/60Hz. [3] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's alternate recomended method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer's operator's manual for appropriate correction factor), without adjusting cabinet airflow balance. Downflow nominal set point was established with I.V.Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492009. [4] May be used with alternate, adjustable height electric stand (maximum bench height of 35 inches). [5] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 63 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492009. Certified for use with power supply of 230V/50Hz and 230V/60Hz. [6] Certified for use with power supply of 220 V / 50 Hz and 220 V / 60 Hz. [7] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal setpoint was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal setpoint of 63 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492009. Certified for use with power supply of 220V/50Hz. [8] Beginning with serial number 1881221 and 1881081 for models 1395 and 1397, respectively. Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 63 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492009. 
Facility : Suzhou, China
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

1384[1] [2]  A2  A  100  110  58  68  N/A  4  10S  36  Armrest I.V. Pole U.V. Light 
1389[1] [2]  A2  A  100  110  58  68  N/A  4  10S  36  Armrest I.V. Pole U.V. Light 
[1] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal setpoint was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal setpoint of 63 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. Certified for use with power supply of 220V/50Hz. [2] Models 1384 and 1389 can be provided with Type F power cord and internal receptacles and identified as 1384F or 1389F; with Type G power cord and internal receptacles and identified as 1384G or 1389G; with Type M power cord and Type F internal receptacles and identified as 1384M or 1389M; or with the standard Type I power cord and internal receptacles. 
Facility : Langenselbold, Germany
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

KS12[1]  A2  A  100  110  70  80  N/A  4  10 10H S 
34  Full Side Windows U.V. Light I.V. Pole Armrest Casters 
KS15[2]  A2  A  100  110  65  75  N/A  5  10 10H S 
35  Full Side Windows I.V. Pole U.V. Light Armrest Casters 
KS18[3]  A2  A  100  110  65  75  N/A  6  10 10H S 
34  Full Side Windows I.V. Pole U.V. Light Casters 
KS9[4]  A2  A  100  110  65  75  N/A  3  10 10H S 
35  Full Side Windows I.V. Pole U.V. Light Armrest Casters 
[1] Beginning with serial number 40292176. Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer's operator's manual for appropriate correction factor, if applicable) without adjusting cabinet airflow balance. This cabinet model was Certifed with a 4 section work surface. This cabinet model was Certified to NSF/ANSI 492007. [2] Beginning with serial number 40349384. Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. This nominal setpoint was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer's operator's manual for appropriate correction factor) without adjusting cabinet airflow balance. This cabinet model was Certified with a 5 section work surface. This cabinet model was Certified to NSF/ANSI 492009. [3] Beginning with serial number 40292176. Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer's operator's manual for appropriate correction factor, if applicable) without adjusting cabinet airflow balance. This cabinet model was Certified with a 6 section work surface. This cabinet model was Certified to NSF/ANSI 492007. [4] Beginning with serial number 40329642. Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. This nominal setpoint was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer's operator's manual for appropriate correction factor) without adjusting cabinet airflow balance. This cabinet model was Certified with a 3 section work surface. This cabinet model was Certified to NSF/ANSI 492009. 
Thermo Fisher Scientific  
401 Millcreek Road  
Marietta, OH 45750  
United States  
7403734763 
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

1310[1] [2]  B2  A  100  110  50  60  734 @ 1.7  4  8S  36  BagIn/ BagOut I.V. Pole U.V. Light 
1360[3] [4]  B2  A  100  110  50  60  1191 @ 2.1  6  8S  36  BagIn/ BagOut I.V. Pole U.V. Light 
[1] Beginning with serial number 130500000. Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted access opening, using the manufacturersupplied probe holder and sash stop (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 55 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. Model 1311 is equivalent to Model 1310, but comes equipped with U.V. lighting (Listed as an acceptable option). [2] Certified for use with alternate power supplies. Model 1315 is equivalent to model 1310, but is rated for use with power supplies of 230V and 50/60 Hz and is equipped with a Chinesecompatible plug. Model 1316 is equivalent, but is rated for use with power supplies of 230V and 50/60 Hz and is equipped with U.V. lighting and a Chinesecompatible plug. Model 1317 is equivalent, but is rated for use with power supplies of 230V and 50/60 Hz and is equipped with a North American 230 Voltcompatible plug. Model 1317F is equivalent, but is rated for use with power supplies of 230V and 50/60 Hz and is equipped with a Type F electrical plug (Schuko). Model 1317G is equivalent, but is rated for use with power supplies of 230V and 50/60 Hz and is equipped with a Type G electrical plug (UKcompatible). Model 1317M is equivalent, but is rated for use with power supplies of 230V and 50/60 Hz and is equipped with a Type M electrical plug (Indiacompatible). Model 1318 is equivalent, but is rated for use with power supplies of 230V and 50/60 Hz and is equipped with U.V. lighting and a North American 230 Voltcompatible plug. Model 1318F is equivalent, but is rated for use with power supplies of 230V and 50/60 Hz and is equipped with a Type F electrical plug (Schuko). Model 1318G is equivalent, but is rated for use with power supplies of 230V and 50/60 Hz and is equipped with a Type G electrical plug (UKcompatible). Model 1318M is equivalent, but is rated for use with power supplies of 230V and 50/60 Hz and is equipped with a Type M electrical plug (Indiacompatible). [3] Beginning with serial number 130500000. Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted access opening, using the manufacturersupplied probe holder and sash stop (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 55 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. Model 1361 is equivalent to Model 1360, but is equipped with U.V. lighting (listed as an acceptable option). [4] Certified for use with alternate power supplies. Model 1362 is equivalent to model 1360, but is rated for use with power supplies of 230V and 50/60 Hz and is equipped with a Chinesecompatible plug. Model 1363 is equivalent, but is rated for use with power supplies of 230V and 50/60 Hz and is equipped with U.V. lighting and a Chinesecompatible plug. Model 1364 is equivalent, but is rated for use with power supplies of 230V and 50/60 Hz and is equipped with a North American 230 Voltcompatible plug. Model 1364F is equivalent, but is rated for use with power supplies of 230V and 50/60 Hz and is equipped with a Type F electrical plug (Schuko). Model 1364G is equivalent, but is rated for use with power supplies of 230V and 50/60 Hz and is equipped with a Type G electrical plug (UKcompatible). Model 1364M is equivalent, but is rated for use with power supplies of 230V and 50/60 Hz and is equipped with a Type M electrical plug (Indiacompatible). Model 1365 is equivalent, but is rated for use with power supplies of 230 V and 50/60 Hz and is equipped with U.V. lighting and a North American 230 Voltcompatible plug. Model 1365F is equivalent, but is rated for use with power supplies of 230V and 50/60 Hz and is equipped with a Type F electrical plug (Schuko). Model 1365G is equivalent, but is rated for use with power supplies of 230V and 50/60 Hz and is equipped with a Type G electrical plug (UKcompatible). Model 1365M is equivalent, but is rated for use with power supplies of 230V and 50/60 Hz and is equipped with a Type M electrical plug (Indiacompatible). 
Number of matching Manufacturers is 16
Number of matching Products is 237
Processing time was 0 seconds