NSF Product and Service Listings
These NSF Official Listings are current as of Tuesday, February 07, 2023 at 12:15 a.m. Eastern Time. Please contact NSF to confirm the status of any Listing, report errors, or make suggestions.
Alert: NSF is concerned about fraudulent downloading and manipulation of website text. Always confirm this information by clicking on the below link for the most accurate information: http://info.nsf.org/Certified/Biosafety/Listings.asp?Company=34080&Standard=049&
Class II (Laminar Flow) Biosafety Cabinetry
Cabinet Style A = Bench Unit With Base or Adjustable Legs Provided B = Bench Unit Without Base or Adjustable Legs Provided C = Console 
Window Type H = Hinged S = Sliding F = Fixed 
Bench Height NA = Not Applicable NP = Not Provided 
CBV = Concurrent Balance Value
Due to a change in nomenclature in NSF/ANSI 49  2002 “Class II (Laminar Flow) Biosafety Cabinetry,” Class II, Type A cabinets are now referred to as Class II, Type A1 and Class II, Types B3 and A/B3 cabinets as Class II, Type A2. Class II, Types B1 and B2 cabinets remain unchanged.
Biosafety cabinet models Listed under NSF/ANSI 49 are approved for use with a power supply of 115V/60Hz, unless otherwise noted. If biosafety cabinets have been approved for use with additional power supplies, the power supply will be indicated in the Official Listing.
Nuaire, Inc.  
2100 Fernbrook Lane  
Plymouth, MN 55447  
United States  
7635531270  
Visit this company's website 
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

NU427400[1] [2] [3]  B1  A  100  110  65  75  282 @ 0.7  4  8S  32  BagIn/ BagOut I.V. Pole U.V. Light 
NU427600[1] [2] [4]  B1  A  100  110  65  75  474 @ 0.9  6  8S  32  I.V. Pole U.V. Light BagIn/ BagOut 
NU540300[5]  A2  A  100  110  55  65  N/A  3  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU540300[5]  A2  A  100  110  55  65  N/A  3  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU540400[5]  A2  A  100  110  55  65  N/A  4  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU540400[5]  A2  A  100  110  55  65  N/A  4  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU540500[5]  A2  A  100  110  57  67  N/A  5  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU540500[5]  A2  A  100  110  57  67  N/A  5  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU540600[5]  A2  A  100  110  55  65  N/A  6  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU540600[5]  A2  A  100  110  55  65  N/A  6  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU543300[5]  A2  A  100  110  55  65  N/A  3  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU543300[5]  A2  A  100  110  55  65  N/A  3  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU543300[5]  A2  A  100  110  55  65  N/A  3  12S  35  I.V. Pole U.V. Light Canopy Connection 
NU543400[5]  A2  A  100  110  55  65  N/A  4  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU543400[5]  A2  A  100  110  55  65  N/A  4  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU543400[5]  A2  A  100  110  55  65  N/A  4  12S  35  I.V. Pole U.V. Light Canopy Connection 
NU543500[5]  A2  A  100  110  57  67  N/A  5  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU543500[5]  A2  A  100  110  57  67  N/A  5  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU543500[5]  A2  A  100  110  57  67  N/A  5  12S  35  I.V. Pole U.V. Light Canopy Connection 
NU543600[5]  A2  A  100  110  55  65  N/A  6  8S  35  U.V. Light I.V. Pole Canopy Connection 
NU543600[5]  A2  A  100  110  55  65  N/A  6  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU543600[5]  A2  A  100  110  55  65  N/A  6  12S  35  I.V. Pole U.V. Light Canopy Connection 
NU545300[6]  A2  A  100  110  55  65  N/A  3  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU545300[6]  A2  A  100  110  55  65  N/A  3  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU545400[7]  A2  A  100  110  55  65  N/A  4  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU545400[7]  A2  A  100  110  55  65  N/A  4  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU545500[8]  A2  A  100  110  55  65  N/A  5  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU545500[8]  A2  A  100  110  55  65  N/A  5  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU545600[9]  A2  A  100  110  55  65  N/A  6  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU545600[9]  A2  A  100  110  55  65  N/A  6  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU560400[10]  B2  A  100  110  55  65  859@1.8  4  8S  36  I.V. Pole U.V. Light 
NU560600[11]  B2  A  100  110  55  35  1373@2.0  6  8S  36  I.V. Pole U.V. Light 
NU565400[10]  B2  A  100  110  55  65  859@1.8  4  8S  36  I.V. Pole U.V. Light 
NU565600[11]  B2  A  100  110  55  65  1373@2.0  6  8S  36  I.V. Pole U.V. Light 
NU677400[12]  A2  A  100  110  55  65  N/A  4  12S  38  Canopy Connection U.V. Light 
NU677500[13]  A2  A  100  110  55  65  N/A  5  12S  38  Canopy Connection U.V. Light 
NU677600[14]  A2  A  100  110  55  65  N/A  6  12S  38  Canopy Connection U.V. Light 
NUFH677500[15]  A2  A  100  110  55  65  N/A  5  12S  38  U.V. Light 
[1] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening for appropriate correction factor, if applicable, without adjusting cabinet airflow balance. [2] Listed model may be suffixed with E. Certified for use with power supplies of 230V/50Hz and 230V/60Hz. [3] Beginning with Series 60 and higher. Downflow nominal setpoint of 70 fpm was established 4 inches above the plane of the window with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492011. This cabinet qualifies with the optional DC motor installed. Listed model may be prefaced with Q, N, L or S (i.e. NUQ427400). [4] Beginning with series 40 and higher. Downflow nominal setpoint of 60 fpm was established 4 inches above the plane of the window with the I.V. Pole and U.V. Light removed. This cabinet qualifies with the optional DC motor installed. Listed model may be prefaced with Q, N, L or S (i.e. NUQ427600). This cabinet model was Certified to NSF/ANSI 492011. [5] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. Listed model may be prefaced with Q, N, L, or S (e.g. NUQ540300). Certified for use with power supplies of 100V/50Hz, 100V/60Hz, 230V/50Hz, and 230V/60Hz. [6] Beginning with serial number 170844102115. Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492014. Listed model may be prefaced with Q, N, L, or S (e.g. NUQ540300). Certified for use with power supplies of 100V/50Hz, 100V/60Hz, 230V/50Hz, and 230V/60Hz. [7] Beginning with serial number 170847102115. Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492020. Listed model may be prefaced with Q, N, L, or S (e.g. NUQ540300). Certified for use with power supplies of 100V/50Hz, 100V/60Hz, 230V/50Hz, and 230V/60Hz. [8] Beginning with serial number 170845102115. Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492014. Listed model may be prefaced with Q, N, L, or S (e.g. NUQ540300). Certified for use with power supplies of 100V/50Hz, 100V/60Hz, 230V/50Hz, and 230V/60Hz. [9] Beginning with serial number 170846102115. Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492014. Listed model may be prefaced with Q, N, L, or S (e.g. NUQ540300). Certified for use with power supplies of 100V/50Hz, 100V/60Hz, 230V/50Hz, and 230V/60Hz. [10] Beginning with serial number 179773041817. Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492016. Certified for use with power supplies of 115V/60Hz, 230V/50Hz and 230V/60Hz. The concurrent balance value was taken with the NSF exhaust system conforming to ASHRAE. However, differences due to the exhaust system, cabinet and instrumentation may be present and product specification information should be obtained from the manufacturer. [11] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492016. Certified for use with power supplies of 115V/60Hz, 230V/50Hz and 230V/60Hz. The concurrent balance value was taken with the NSF exhaust system conforming to ASHRAE. However, differences due to the exhaust system, cabinet and instrumentation may be present and product specification information should be obtained from the manufacturer. [12] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492011. NuAire canopy model NU911402 was evaluated as part of the NSF Certification testing procedures. [13] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492011. NuAire canopy model NU911502 was evaluated as a part of the NSF Certification testing procedures. [14] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492011. NuAire canopy model NU911602 was evaluated as part of the NSF Certification testing procedures. [15] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. NuAire canopy model NU911502 was evaluated as a part of the NSF Certification testing procedures. NUFH677500 is a modified version of NU677500, including feed hoppers. 
Facility : Shanghai, China
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

NU543300[1]  A2  A  100  110  55  65  N/A  3  10S  35  Canopy Connection I.V. Pole U.V. Light 
NU543400[1]  A2  A  100  110  55  65  N/A  4  10S  35  Canopy Connection I.V. Pole U.V. Light 
NU543500[2]  A2  A  100  110  57  67  N/A  5  10S  35  Canopy Connection I.V. Pole U.V. Light 
NU543600[1]  A2  A  100  110  55  65  N/A  6  10S  35  Canopy Connection I.V. Pole U.V. Light 
[1] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the I.V. pole and U.V. light removed. This cabinet model was Certified to NSF 492012. Listed model may be prefaced with Q, N, L, or S (e.g. NUQ540300). Listed model may be suffixed with J, K, or S (e.g. NU543400S). Certified for use with power supplies of 100V/50Hz, 100V/60Hz, 230V/50Hz, and 230V/60Hz. [2] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 62 fpm was established 4 inches above the bottom of the sash with the I.V. pole and U.V. light removed. This cabinet model was Certified to NSF 492012. Listed model may be prefaced with Q, N, L, or S (e.g. NUQ540300). Listed model may be suffixed with J, K, or S (e.g. NU543400S). Certified for use with power supplies of 100V/50Hz, 100V/60Hz, 230V/50Hz, and 230V/60Hz. 
Number of matching Manufacturers is 1
Number of matching Products is 42
Processing time was 0 seconds