NSF Product and Service Listings
These NSF Official Listings are current as of Thursday, October 21, 2021 at 12:15 a.m. Eastern Time. Please contact NSF to confirm the status of any Listing, report errors, or make suggestions.
Alert: NSF is concerned about fraudulent downloading and manipulation of website text. Always confirm this information by clicking on the below link for the most accurate information: http://info.nsf.org/Certified/Biosafety/Listings.asp?TradeName=+&
Class II (Laminar Flow) Biosafety Cabinetry
Cabinet Style A = Bench Unit With Base or Adjustable Legs Provided B = Bench Unit Without Base or Adjustable Legs Provided C = Console 
Window Type H = Hinged S = Sliding F = Fixed 
Bench Height NA = Not Applicable NP = Not Provided 
CBV = Concurrent Balance Value
Due to a change in nomenclature in NSF/ANSI 49  2002 “Class II (Laminar Flow) Biosafety Cabinetry,” Class II, Type A cabinets are now referred to as Class II, Type A1 and Class II, Types B3 and A/B3 cabinets as Class II, Type A2. Class II, Types B1 and B2 cabinets remain unchanged.
Biosafety cabinet models Listed under NSF/ANSI 49 are approved for use with a power supply of 115V/60Hz, unless otherwise noted. If biosafety cabinets have been approved for use with additional power supplies, the power supply will be indicated in the Official Listing.
Across International LLC  
111 Dorsa Avenue  
Livingston, NJ 07039  
United States  
8889880899  
Visit this company's website 
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

BC3F[1]  A2  A  100  110  60  70  N/A  3  10S  27  Canopy Connection I.V. Pole U.V. Light 
BC4F[2]  A2  A  100  110  60  70  N/A  4  10S  27  I.V. Pole U.V. Light Canopy Connection 
BC4FB2[1]  B2  A  100  110  60  70  949 @ 2.1  4  8S  27  I.V. Pole U.V. Light 
BC6F[3]  A2  A  100  110  60  70  N/A  6  10S  27  I.V. Pole Canopy Connection U.V. Light 
[1] Beginning with serial number 3FA18060001. Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer's operator's manual for appropriate correction factor) without adjusting cabinet airflow balance. Downflow nominal set point of 65 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492016. Biobase canopy model CY3FA2I was evaluated as a part of the NSF Certification testing procedures. Certified for use with power supplies of 115V/60Hz, 230V/50Hz and 230V/60Hz. [2] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 65 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492014. Certified for use with power supplies of 115V/60Hz, 230V/50Hz and 230V/60Hz. [3] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 65 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492014. Certified for use with power supplies of 115V/60Hz, 230V/50Hz and 230V/60Hz. 
Allentown, LLC  
Route 526  
Allentown, NJ 085010698  
United States  
6092597951  
Visit this company's website 
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

39241[1] [2]  A2  A  100  110  50  60  N/A  4  10S  36  Canopy Connection I.V. Pole U.V. Light 
39242xxx1 Safety Cab.[1] [3]  A2  A  100  110  55  65  N/A  4  12S  36  Canopy Connection I.V. Pole U.V. Light 
39248[1] [2]  A2  A  100  110  50  60  N/A  4  8S  36  Canopy Connection I.V. Pole U.V. Light 
39251[1] [2]  A2  A  100  110  50  60  N/A  5  10S  36  Canopy Connection I.V. Pole U.V. Light 
39252xxx1 Safety Cab.[1] [4]  A2  A  100  110  55  65  N/A  5  12S  36  Canopy Connection I.V. Pole U.V. Light 
39258[1] [2]  A2  A  100  110  50  60  N/A  5  8S  36  Canopy Connection I.V. Pole U.V. Light 
39261[1] [2]  A2  A  100  110  50  60  N/A  6  10S  36  Canopy Connection I.V. Pole U.V. Light 
39262xxx1 Safety Cab.[1] [3]  A2  A  100  110  55  65  N/A  6  12S  36  Canopy Connection I.V. Pole U.V. Light 
39268[1] [2]  A2  A  100  110  50  60  N/A  6  8S  36  Canopy Connection I.V. Pole U.V. Light 
[1] Certified for use with power supplies of 100V/50Hz, 100V/60Hz, 230V/50Hz, and 230V/60Hz. [2] Inflow nominal set point of 105 fpm was established with a direct airflow reading measurement. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted access opening, using the manufacturersupplied probe holder and sash stop (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 55 fpm was established with I.V. pole and U.V. light removed. This cabinet model was Certified to NSF/ANSI 492012. This cabinet model is Certified for use with optional sidewall VacuPassTM Cord and Cable Portal. [3] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted access opening, using the manufacturersupplied probe holder and sash stop (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 55 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492018. This cabinet model is Certified for use with optional sidewall VacuPassTM Cord & Cable Portal. [4] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted access opening, using the manufacturersupplied probe holder and sash stop (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 60 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492018. This cabinet model is Certified for use with optional sidewall VacuPassTM Cord & Cable Portal. 
Baker Company, Inc. (The)  
175 Gatehouse Road  
Sanford, ME 04073  
United States  
8009922537  
2073248773  
Visit this company's website 
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

BCG401[1]  B2  C  100  110  35  45  664 @ 1.6  4  8S  35  I.V. Pole U.V. Light Casters 
BCG601[1]  B2  C  100  110  35  45  993 @ 1.6  6  8S  35  I.V. Pole U.V. Light 
SG404[2] [3] [4] [5]  A2  A  100  110  47  57 42  52 
N/A  4  8S  37  I.V. Pole U.V. Light Casters Canopy Connection 
SG404[2] [3] [4] [5]  A2  A  100  110  47  57 42  52 
N/A  4  10S  37  I.V. Pole U.V. Light Casters Canopy Connection 
SG404[2] [3] [4] [5]  A2  A  100  110  47  57 42  52 
N/A  4  12S  37  I.V. Pole U.V. Light Casters Canopy Connection 
SG504[2] [3] [5] [6]  A2  A  100  110  45  55 42  52 
N/A  5  8S  37  I.V. Pole U.V. Light Casters Canopy Connection 
SG504[2] [3] [5] [6]  A2  A  100  110  45  55 42  52 
N/A  5  10S  37  I.V. Pole U.V. Light Casters Canopy Connection 
SG504[2] [3] [5] [7]  A2  A  100  110  45  55 40  50 
N/A  5  12S  37  I.V. Pole U.V. Light Casters Canopy Connection 
SG604[2] [3] [5] [8]  A2  A  100  110  48  58 42  52 
N/A  6  8S  37  Casters Canopy Connection I.V. Pole U.V. Light 
SG604[2] [3] [5] [8]  A2  A  100  110  48  58 42  52 
N/A  6  10S  37  Casters Canopy Connection I.V. Pole U.V. Light 
SG604[2] [3] [5] [9]  A2  A  100  110  42  52 36  46 
N/A  6  12S  37  Canopy Connection Casters I.V. Pole U.V. Light 
[1] Beginning with serial number 103539 for the 4foot model and serial number 103706 for the 6foot model. Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate calculated method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer's operator's manual for appropriate correction factor), without adjusting cabinet airflow balance. Downflow nominal set point was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492010a. This cabinet qualifies only with the armrest installed. This cabinet is approved with additional petcocks on the sidewalls and a low static pressure exhaust plenum. Cabinet models BCG401 INT and BCG601 INT are Certified for alternate motor/blower combinations and input power supply. [2] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate calculated method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer's operator's manual for appropriate correction factor, if applicable). [3] Baker model designations SG404AUS and SG604AUS are Certified to NSF/ANSI 492012 for operation at 220V/50Hz with an 8inch window opening. Baker model designations SG404INT, SG504INT, and SG604INT are Certified to NSF/ANSI 492012 for operation at either 220V/50Hz or 220V/60Hz. [4] Beginning with serial number 110235. The acceptable nonuniform downflow ranges listed above begin with zone 1 (4757 fpm) at the rear of the work area (compromised of lateral rear and center rows) and end with zone 2 (4252 fpm) (comprised of front lateral row) behind the front window enclosing the work area. Downflow nominal set points were established with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. This cabinet qualifies only with the armrest installed. FlexAir Canopy connection 333A745 is approved for use with this model. [5] The maximum height for the bench stand with casters is 30 inches. The maximum height for the bench stand with feet is 37 inches. [6] Beginning with serial number 110795. The acceptable nonuniform downflow ranges listed above begin with zone 1 (4555 fpm) at the rear of the work area (compromised of lateral rear and center rows) and end with zone 2 (4252 fpm) (comprised of front lateral row) behind the front window enclosing the work area. Downflow nominal set points were established with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. This cabinet qualifies only with the armrest installed. FlexAir Canopy connection 336A745 is approved for use with this model. [7] Beginning with serial number 110795. The acceptable nonuniform downflow ranges listed above begin with zone 1 (4555 fpm) at the rear of the work area (compromised of lateral rear and center rows) and end with zone 2 (4050 fpm) (comprised of front lateral row) behind the front window enclosing the work area. Downflow nominal set points were established with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. This cabinet qualifies only with the armrest installed. FlexAir Canopy connection 336A745 is approved for use with this model. [8] Beginning with serial number 110555. The acceptable nonuniform downflow ranges listed above begin with zone 1 (4858 fpm) at the rear of the work area (compromised of lateral rear and center rows) and end with zone 2 (4252 fpm) (comprised of front lateral row) behind the front window enclosing the work area. Downflow nominal set points were established with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. This cabinet qualifies only with the armrest installed. FlexAir Canopy connection 336A745 is approved for use with this model. [9] Beginning with serial number 110555. The acceptable nonuniform downflow ranges listed above begin with zone 1 (4252 fpm) at the rear of the work area (compromised of lateral rear and center rows) and end with zone 2 (3646 fpm) (comprised of front lateral row) behind the front window enclosing the work area. Downflow nominal set points were established with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. This cabinet qualifies only with the armrest installed. FlexAir Canopy connection 336A745 is approved for use with this model. 
CHC Lab  
5201 Youngsandong  
Yusunggu  
Daejeon 305500  
Korea, Republic of  
82 42 933 0036  
Visit this company's website 
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

CHC777A204[1] [3]  A2  A  100  110  65  75  N/A  4  8S  33  Casters I.V. Pole U.V. Light 
CHC777A206[1] [4]  A2  A  100  110  65  75  N/A  6  8S  33  Casters I.V. Pole U.V. Light 
CHC888B204[2] [4]  B2  A  100  110  60  70  818 @ 2.4  4  8S  33  I.V. Pole U.V. Light 
CHC888B206[4] [5]  B2  A  100  110  55  65  1231@3.16  6  8S  32  I.V. Pole U.V. Light 
[1] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's alternate recomemded method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer's operator's manual for appropriate correction factor), without adjusting cabinet airflow balance. Downflow nominal set point was established with I.V.Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492007. This cabinet qualifies only with the armrest installed. [2] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's alternate recomemded method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer's operator's manual for appropriate correction factor), without adjusting cabinet airflow balance. Downflow nominal set point was established with I.V.Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492009. [3] Certified for use with power supplies of 115V/60 Hz, 220V/50 Hz, and 220V/60 Hz. [4] Certified for use with power supplies of 115V/60 Hz and 220V/60 Hz. [5] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's alternate recommended method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer's operator's manual for appropriate correction factor), without adjusting cabinet airflow balance. Downflow nominal set point was established with I.V.Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492010. This cabinet qualifies only with the armrest installed. 
Chung Fu Technical Development Co., Ltd.  
7F, No. 687, Mintsu East Road  
Taipei 105  
Taiwan  
886 2 2600 1761  
Visit this company's website 
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

AHA133AAB[1]  A2  A  100  110  54  64  N/A  4  10S  28  I.V. Pole U.V. Light 
AHA133ABB[1] [2]  A2  A  100  110  54  64  N/A  4  10S  28  I.V. Pole U.V. Light 
AHA133ACB[1] [3]  A2  A  100  110  54  64  N/A  4  10S  28  I.V. Pole U.V. Light 
AHA133CAB[1]  A2  A  100  110  54  64  N/A  4  10S  28  I.V. Pole U.V. Light 
AHA133CBB[1] [2]  A2  A  100  110  54  64  N/A  4  10S  28  I.V. Pole U.V. Light 
AHA133CCB[1] [3]  A2  A  100  110  54  64  N/A  4  10S  28  I.V. Pole U.V. Light 
[1] Beginning with serial number AHA1608060. Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal set point of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 59 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492014. Chung Fu Technical Development Co., Ltd. Models AHA133AAB, AHA133ABB, AHA133ACB, AHA133CAB, AHA133CBB and AHA133CCB are marketed by Air Science, LLC, USA as Air Science Models ASAHA133AAB, ASAHA133ABB, ASAHA133ACB, ASAHA133CAB, ASAHA133CBB and ASAHA133CCB, respectively. [2] Certified for use with power supplies of 230V/50Hz and 230V/60Hz. [3] Certified for use with power supplies of 100V/50Hz and 100V/60Hz. 
Design Filtration Microzone Inc.  
86 Harry Douglas Drive  
Ottawa, ON K2S 2C7  
Canada  
6138318318  
Visit this company's website 
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

BK34[1]  A2  A  100  110  60  70  N/A  4  10S  27  Canopy Connection I.V. Pole U.V. Light 
BK36[2]  A2  A  100  110  60  70  N/A  6  10S  27  I.V. Pole U.V. Light Canopy Connection 
BK34[3] [4]  A2  A  100  110  60  70  N/A  4  8S  27  I.V. Pole U.V. Light 
[1] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal set point of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal set point of 65 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492014. Certified for use with power supplies of 115V/60Hz, 230V/50Hz and 230V/60Hz. [2] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 65 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492014. Certified for use with power supplies of 115V/60Hz, 230V/50Hz and 230V/60Hz. [3] Approved for alternate power modes of 220V/50Hz and 220V/60Hz. [4] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal setpoint was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal setpoint of 65 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. 
Esco Micro Pte Ltd  
21 Changi South Street 1  
486777  
Singapore  
65 6542 0833  
Visit this company's website 
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

AC23S8NS[1] [2] [3]  A2  A  100  110  55  65  N/A  3  8S  35  I.V. Pole U.V. Light Canopy Connection 
AC23S9NS[1]  A2  A  100  100  55  65  N/A  3  8S  35  I.V. Pole U.V. Light Canopy Connection 
AC24N7[4] [5]  A2  A  100  110  55  65  N/A  4  8S  36  I.V. Pole U.V. Light 
AC24S8NS[2] [3] [4]  A2  A  100  110  55  65  N/A  4  8S  36  I.V. Pole U.V. Light 
AC24S9NS[4]  A2  A  100  110  55  65  N/A  4  8S  36  I.V. Pole U.V. Light 
AC24Y7[4] [5]  A2  A  100  110  55  65  N/A  4  8S  36  I.V. Pole U.V. Light 
AC25S8NS[2] [3] [6]  A2  A  100  110  55  65  N/A  5  8S  35  I.V. Pole U.V. Light 
AC25S9NS[6]  A2  A  100  110  55  65  N/A  5  8S  35  I.V. Pole U.V. Light 
AC26N7[5] [7]  A2  A  100  110  55  65  N/A  6  8S  36  I.V. Pole U.V. Light 
AC26S8NS[2] [3] [7]  A2  A  100  110  55  65  N/A  6  8S  36  I.V. Pole U.V. Light 
AC26S9NS[7]  A2  A  100  110  55  65  N/A  6  8S  36  I.V. Pole U.V. Light 
AC26Y7[5] [7]  A2  A  100  110  55  65  N/A  6  8S  36  I.V. Pole U.V. Light 
AR23S8[1] [2] [3]  A2  A  100  110  55  65  N/A  3  8S  35  I.V. Pole U.V. Light Canopy Connection 
AR23S9[1]  A2  A  100  110  55  65  N/A  3  8S  35  I.V. Pole U.V. Light Canopy Connection 
AR24S9[4]  A2  A  100  110  55  65  N/A  4  8S  36  I.V. Pole U.V. Light 
AR25S8[2] [3] [6]  A2  A  100  110  55  65  N/A  5  8S  35  I.V. Pole U.V. Light 
AR25S9[6]  A2  A  100  110  55  65  N/A  5  8S  35  I.V. Pole U.V. Light 
AR26S9[7]  A2  A  100  110  55  65  N/A  6  8S  36  I.V. Pole U.V. Light 
BC24S7[4] [5]  A2  A  100  100  55  65  N/A  4  8S  36  I.V. Pole U.V. Light 
BC26S7[5] [7]  A2  A  100  110  55  65  N/A  6  8S  36  I.V. Pole U.V. Light 
LA24A1E[2] [8]  A2  A  100  110  65  75  N/A  4  9S  34  I.V. Pole U.V. Light 
LA24A2E[8]  A2  A  100  110  65  75  N/A  4  9S  34  I.V. Pole U.V. Light 
LA24A3E[3] [8]  A2  A  100  110  65  75  N/A  4  9S  34  I.V. Pole U.V. Light 
LA25A1E[2] [8]  A2  A  100  110  65  75  N/A  5  9S  36  I.V. Pole U.V. Light 
LA25A2E[8]  A2  A  100  110  65  75  N/A  5  9S  36  I.V. Pole U.V. Light 
LA25A3E[3] [8]  A2  A  100  110  65  75  N/A  5  9S  36  I.V. Pole U.V. Light 
LA26A1E[2] [8]  A2  A  100  110  60  70  N/A  6  8S  35  I.V. Pole U.V. Light 
LA26A2E[8]  A2  A  100  110  60  70  N/A  6  8S  35  I.V. Pole U.V. Light 
LA26A3E[3] [8]  A2  A  100  110  60  70  N/A  6  8S  35  I.V. Pole U.V. Light 
LB24B1E[2] [9]  B2  A  100  110  55  65  803 @ 1.9  4  8S  36  I.V. Pole U.V. Light 
LB24B2E[9]  B2  A  100  110  55  65  803 @ 1.9  4  8S  36  I.V. Pole U.V. Light 
LB24B3E[3] [9]  B2  A  100  110  55  65  803 @ 1.9  4  8S  36  I.V. Pole U.V. Light 
LB25B1E[2] [9] [10]  B2  A  100  110  55  65  1025 @ 1.8  5  8S  35  I.V. Pole U.V. Light 
LB25B2E[9] [10]  B2  A  100  110  55  65  1025 @ 1.8  5  8S  35  I.V. Pole U.V. Light 
LB25B3E[3] [9] [10]  B2  A  100  110  55  65  1025 @ 1.8  5  8S  35  I.V. Pole U.V. Light 
LB26B1E[2] [9]  B2  A  100  110  55  65  1269 @ 2.1  6  8S  36  I.V. Pole U.V. Light 
LB26B2E[9]  B2  A  100  110  55  65  1269 @ 2.1  6  8S  36  I.V. Pole U.V. Light 
LB26B3E[3] [9]  B2  A  100  110  55  65  1269 @ 2.1  6  8S  36  I.V. Pole U.V. Light 
LR24S1E[2] [8]  A2  A  100  110  65  75  N/A  4  9S  34  I.V. Pole U.V. Light 
LR24S2E[8]  A2  A  100  110  65  75  N/A  4  9S  34  I.V. Pole U.V. Light 
LR24S3E[3] [8]  A2  A  100  110  65  75  N/A  4  9S  34  I.V. Pole U.V. Light 
LR25S1E[2] [8]  A2  A  100  110  65  75  N/A  5  9S  36  I.V. Pole U.V. Light 
LR25S2E[8]  A2  A  100  110  65  75  N/A  5  9S  36  I.V. Pole U.V. Light 
LR25S3E[3] [8]  A2  A  100  110  65  75  N/A  5  9S  36  I.V. Pole U.V. Light 
LR26S1E[2] [8]  A2  A  100  110  60  70  N/A  6  8S  35  I.V. Pole U.V. Light 
LR26S2E[8]  A2  A  100  110  60  70  N/A  6  8S  35  I.V. Pole U.V. Light 
LR26S3E[3] [8]  A2  A  100  110  60  70  N/A  6  8S  35  I.V. Pole U.V. Light 
[1] Beginning with serial number 2015  94341. Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 60 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492014. [2] Certified for use with a power supply of 230V/50Hz. [3] Certified for use with a power supply of 230V/60Hz. [4] Beginning with serial number 2014  89128. Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 60 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. [5] Approved for alternate power modes of 100V/50Hz and 100V/60Hz. [6] Beginning with serial number 2015  94651. Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 60 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492014. [7] Beginning with serial number 2014  84641. Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 60 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. [8] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 70 fpm was established with I.V. Pole and U.V. Light removed.This cabinet model was Certified to NSF/ANSI 492011. [9] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 60 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. [10] Certified for use as benchtop model or with SPL5B0 stand only. 
JEIO TECH Co., Ltd.  
153 Techno 2ro Yuseonggu  
Daejeon 34025  
Korea, Republic of  
82 42 933 4296  
Visit this company's website 
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

JB12[1]  A2  A  100  110  56  66  N/A  4  8.27S  28  None 
[1] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (80 mm high) access opening (consult manufacturer's operator's manual for appropriate correction factor) without adjusting cabinet airflow balance. This cabinet model was Certified to NSF/ANSI 492018 for use with an input power supply of 230V/60Hz only. 
Jinan Biobase Biotech Co., Ltd.  
North side of Jiwang Road, Mingshui  
Economic Dev. Zone, Zhangqiu Jinan  
Shandong Province 250200  
China  
86 531 5867 3159  
Visit this company's website 
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

BSC3FA2GL[1]  A2  A  100  110  60  70  N/A  3  10S  27  Canopy Connection I.V. Pole U.V. Light 
BSC3FA2NA[1]  A2  A  100  110  60  70  N/A  3  10S  27  Canopy Connection I.V. Pole U.V. Light 
BSC4FA2[2] [3]  A2  A  100  110  60  70  N/A  4  8S  27  I.V. Pole U.V. Light 
BSC4FA2GL[4]  A2  A  100  110  60  70  N/A  4  10S  27  I.V. Pole U.V. Light Canopy Connection 
BSC4FA2NA[4]  A2  A  100  110  60  70  N/A  4  10S  27  I.V. Pole U.V. Light Canopy Connection 
BSC4FB2GL[5]  B2  A  100  110  60  70  949 @ 2.1  4  8S  27  I.V. Pole U.V. Light 
BSC4FB2NA[5]  B2  A  100  110  60  70  949 @ 2.1  4  8S  27  I.V. Pole U.V. Light 
BSC6FA2GL[6]  A2  A  100  110  60  70  N/A  6  10S  27  I.V. Pole Canopy Connection U.V. Light 
BSC6FA2NA[6]  A2  A  100  110  60  70  N/A  6  10S  27  I.V. Pole Canopy Connection U.V. Light 
[1] Beginning with serial number 3FA18060001. Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer's operator's manual for appropriate correction factor) without adjusting cabinet airflow balance. Downflow nominal set point of 65 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492016. Biobase canopy model CY3FA2I was evaluated as a part of the NSF Certification testing procedures. Certified for use with power supplies of 115V/60Hz, 230V/50Hz and 230V/60Hz. [2] Approved for alternate power modes of 220V/50Hz and 220V/60Hz. [3] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal set point of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal set point of 65 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. [4] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 65 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492014. Certified for use with power supplies of 115V/60Hz, 230V/50Hz and 230V/60Hz. [5] Beginning with serial number 4FB18050001. Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer's operator's manual for appropriate correction factor) without adjusting cabinet airflow balance. Downflow nominal set point of 65 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492016. Certified for use with power supplies of 115V/60Hz, 230V/50Hz and 230V/60Hz. [6] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 65 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492014. Certified for use with power supplies of 115V/60Hz, 230V/50Hz and 230V/60Hz. 
Kewaunee Scientific Corporation  
2700 West Front Street  
Statesville, NC 28677  
United States  
7048737202  
Visit this company's website 
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

INT11002[1] [2]  A2  A  100  110  60  70  N/A  3  10S  35  I.V. Pole U.V. Light 
INT14002[2] [3]  A2  A  100  110  50  60  N/A  4  10S  35  I.V. Pole U.V. Light 
INT17002[2] [4]  A2  A  100  110  55  65  N/A  5  10S  35  I.V. Pole U.V. Light 
INT20002[2] [3]  A2  A  100  110  50  60  N/A  6  10S  35  I.V. Pole U.V. Light 
INTB14002[5] [6]  B2  A  100  110  55  65  937@2.4  4  8S  36  I.V. Pole U.V. Light 
INTB20002[5] [6]  B2  A  100  110  55  65  1416@2.0  6  8S  36  I.V. Pole U.V. Light 
[1] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 65 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model qualifies only with the armrest installed. This cabinet model was Certified to NSF/ANSI 492014. [2] Also Certified for use with power supplies of 230V/50Hz and 230V/60Hz. [3] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. This nominal setpoint was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer for appropriate correction factor, if applicable) without adjusting cabinet airflow balance. Downflow nominal setpoint of 55 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model qualifies only with the armrest installed. This cabinet model was Certified to NSF/ANSI 492014. [4] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 60 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model qualifies only with the armrest installed. This cabinet model was Certified to NSF/ANSI 492014. [5] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 60 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492016. [6] Also Certified for use with power supplies of 230V/50Hz and 230V/60Hz. Model numbers may be followed by 230 suffix. 
Facility : Karnataka, India
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

INT11002[1] [2]  A2  A  100  110  60  70  N/A  3  10S  35  I.V. Pole U.V. Light 
INT14002[2] [3]  A2  A  100  110  50  60  N/A  4  10S  35  I.V. Pole U.V. Light 
INT17002[2] [4]  A2  A  100  110  55  65  N/A  5  10S  35  I.V. Pole U.V. Light 
INT20002[2] [3]  A2  A  100  110  50  60  N/A  6  10S  35  U.V. Light I.V. Pole 
INTB14002[5] [6]  B2  A  100  110  55  65  937@2.4  4  8S  36  I.V. Pole U.V. Light 
INTB20002[5] [6]  B2  A  100  110  55  65  1416@2.0  6  8S  36  I.V. Pole U.V. Light 
[1] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 65 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model qualifies only with the armrest installed. This cabinet model was Certified to NSF/ANSI 492014. [2] Certified for use with power supplies of 230V/50Hz and 230V/60Hz. [3] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. This nominal setpoint was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening consult manufacturer for appropriate correction factor, if applicable) without adjusting cabinet airflow balance. Downflow nominal setpoint of 55 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model qualifies only with the armrest installed. This cabinet model was Certified to NSF/ANSI 492014. [4] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 60 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model qualifies only with the armrest installed. This cabinet model was Certified to NSF/ANSI 492014. [5] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 60 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492016. [6] Also Certified for use with power supplies of 230V/50Hz and 230V/60Hz. Model numbers may be followed by 230 suffix. 
Labconco Corporation  
8845 Prospect  
Kansas City, MO 64132  
United States  
8163338811 X771  
Visit this company's website 
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

30231xxx1 Logic+[1] [2]  A2  A  100  110  50  60  N/A  3  10S  36  I.V. Pole U.V. Light Canopy Connection 
30238xxx1 Logic+[1] [3]  A2  A  100  110  50  60  N/A  3  8S  36  I.V. Pole U.V. Light Canopy Connection 
30241xxx1 Logic+[1] [3]  A2  A  100  110  50  60  N/A  4  10S  36  Canopy Connection I.V. Pole U.V. Light 
30242xxx1 Logic+[3] [4]  A2  A  100  110  50  60  N/A  4  12S  36  Canopy Connection I.V. Pole U.V. Light 
30248xxx1 Logic+[1] [3]  A2  A  100  110  50  60  N/A  4  8S  36  Canopy Connection I.V. Pole U.V. Light 
30251xxx1 Logic+[1] [3]  A2  A  100  110  50  60  N/A  5  10S  36  I.V. Pole U.V. Light Canopy Connection 
30252xxx1 Logic+[3] [5]  A2  A  100  110  55  65  N/A  5  12S  36  Canopy Connection I.V. Pole U.V. Light 
30258xxx1 Logic+[1] [3]  A2  A  100  110  50  60  N/A  5  8S  36  Canopy Connection I.V. Pole U.V. Light 
30261xxx1 Logic+[1] [3]  A2  A  100  110  50  60  N/A  6  10S  36  Canopy Connection I.V. Pole U.V. Light 
30262xxx1 Logic+[3] [4]  A2  A  100  110  50  60  N/A  6  12S  36  Canopy Connection I.V. Pole U.V. Light 
30268xxx1 Logic+[1] [3]  A2  A  100  110  50  60  N/A  6  8S  36  Canopy Connection I.V. Pole U.V. Light 
30348xxx1 Logic+[1] [3]  B2  A  100  110  50  60  852 @ 1.8  4  8S  36  BagIn/ BagOut I.V. Pole U.V. Light 
30368xxx1 Logic+[1] [3]  B2  A  100  110  50  60  1265 @ 2.2  6  8S  36  I.V. Pole U.V. Light BagIn/ BagOut 
30448xxx1 Axiom[3] [6]  C1  A  100  110  60  70  N/A  4  8S  36  I.V. Pole U.V. Light BagIn/ BagOut 
30441xxx1 Axiom[3] [6]  C1  A  100  110  60  70  N/A  4  10S  36  I.V. Pole U.V. Light BagIn/ BagOut 
30468xxx1 Axiom[3] [4]  C1  A  100  110  50  60  N/A  6  8S  36  I.V. Pole U.V. Light BagIn/ BagOut 
30461xxx1 Axiom[3] [6]  C1  A  100  110  60  70  N/A  6  10S  36  I.V. Pole U.V. Light BagIn/ BagOut 
31231xxx1 PuriCare[1] [2]  A2  A  100  110  50  60  N/A  3  10S  36  I.V. Pole U.V. Light Canopy Connection 
31241xxx1 PuriCare[1] [3]  A2  A  100  110  50  60  N/A  4  10S  36  Canopy Connection I.V. Pole U.V. Light 
31242xxx1 PuriCare[3] [4]  A2  A  100  110  50  60  N/A  4  12S  36  I.V. Pole U.V. Light Canopy Connection 
31251xxx1 PuriCare[1] [3]  A2  A  100  110  50  60  N/A  5  10S  36  Canopy Connection I.V. Pole U.V. Light 
31252xxx1 PuriCare[3] [7]  A2  A  100  110  55  65  N/A  5  12S  36  Canopy Connection I.V. Pole U.V. Light 
31261xxx1 PuriCare[1] [3]  A2  A  100  110  50  60  N/A  6  10S  36  Canopy Connection I.V. Pole U.V. Light 
31262xxx1 PuriCare[3] [4]  A2  A  100  110  50  60  N/A  6  12S  36  Canopy Connection I.V. Pole U.V. Light 
[1] Beginning with serial number 190500000. Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted access opening, using the manufacturersupplied probe holder and sash stop (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 55 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492016. This cabinet model is Certified for use with optional sidewall Vacu PassTM Cord & Cable Portal. [2] Certified for use with power supplies of 230V/50 Hz and 230V/60 Hz. [3] Certified for use with power supplies of 100V/50 Hz, 100V/60 Hz, 230V/50 Hz, and 230V/60 Hz. [4] Beginning with serial number 151200002. Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted access opening, using the manufacturersupplied probe holder and sash stop (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 60 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492014. This cabinet model is Certified for use with optional sidewall VacuPassTM Cord & Cable Portal. [5] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted access opening, using the manufacturersupplied probe holder and sash stop (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 55 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492018. This cabinet model is Certified for use with optional sidewall VacuPassTM Cord & Cable Portal. [6] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted access opening, using the manufacturersupplied probe holder and sash stop (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 60 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492018. This cabinet model is Certified for use with optional sidewall VacuPassTM Cord & Cable Portal. [7] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted access opening, using the manufacturersupplied probe holder and sash stop (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 65 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492018. This cabinet model is Certified for use with optional sidewall VacuPassTM Cord & Cable Portal. 
NOTE: Bench stand models 3730300, 3730310, 3401003, 3780200 and 3780100 may be used with cabinets in the 30231xxx1/30238xxx1 Logic+ series. Bench stand models 3730400, 3730410, 3401004, 3780201 and 3780101 may be used with cabinets in the 30241xxx1/30248xxx1/30242 Logic+ series and the corresponding PuriCare series. Bench stand models 3730500, 3730510, 3401005, 3780203, 3780106, 3780107, 3780312, and 3780316 may be used with cabinets in the 30251xxx1/30258xxx1 Logic+ series. Bench stand models 3730600, 3630610, 3401006, 3780202 and 3780102 may be used with cabinets in the 30261xxx1/30268xxx1/30262 Logic+ series and the corresponding PuriCare series. Bench stand models 3730400 and 34103003 may also be used with the 30348xxx1 Logic+ Series. Bench stand models 3730600 and 34103006 may also be used with the 30368xxx1 Logic+ Series. All bench stand models beginning with 378 may be used with or without the optional wheel kit (PN# 3784000).
MEDFUTURE LLC  
3231 Osgood Common  
Freemont, CA 94539  
United States  
6262834740  
Visit this company's website 
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

BIIA24F[1] [2]  A2  A  100  110  60  70  N/A  4  8S  27  I.V. Pole U.V. Light Canopy Connection 
BIIA24FGL[3]  A2  A  100  110  60  70  N/A  4  10S  27  I.V. Pole U.V. Light Canopy Connection 
BIIA24FNA[3]  A2  A  100  110  60  70  N/A  4  10S  27  I.V. Pole U.V. Light Canopy Connection 
BIIA26FGL[4]  A2  A  100  110  60  70  N/A  6  10S  27  I.V. Pole U.V. Light Canopy Connection 
BIIA26FNA[4]  A2  A  100  110  60  70  N/A  6  10S  27  I.V. Pole U.V. Light Canopy Connection 
[1] Approved for alternate power modes of 220V/50Hz and 220V/60Hz. [2] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal set point of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal set point of 65 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. [3] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 65 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492014. Certified for use with power supplies of 115V/60Hz, 230V/50Hz and 230V/60Hz. [4] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 65 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492014. Certified for use with power supplies of 115V/60Hz, 230V/50Hz and 230V/60Hz. 
Nuaire, Inc.  
2100 Fernbrook Lane  
Plymouth, MN 55447  
United States  
7635531270  
Visit this company's website 
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

NU427400[1] [2] [3]  B1  A  100  110  65  75  282 @ 0.7  4  8S  32  BagIn/ BagOut I.V. Pole U.V. Light 
NU427600[1] [2] [4]  B1  A  100  110  65  75  474 @ 0.9  6  8S  32  I.V. Pole U.V. Light BagIn/ BagOut 
NU430400[1] [2] [5] [6]  B2  A  100  110  55  65  785 @ 1.7  4  8S  32  BagIn/ BagOut I.V. Pole U.V. Light 
NU430600[1] [2] [5] [7]  B2  A  100  110  55  65  1250 @ 1.8  6  8S  32  BagIn/ BagOut I.V. Pole U.V. Light 
NU540300[8]  A2  A  100  110  55  65  N/A  3  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU540300[8]  A2  A  100  110  55  65  N/A  3  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU540400[8]  A2  A  100  110  55  65  N/A  4  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU540400[8]  A2  A  100  110  55  65  N/A  4  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU540500[8]  A2  A  100  110  57  67  N/A  5  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU540500[8]  A2  A  100  110  57  67  N/A  5  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU540600[8]  A2  A  100  110  55  65  N/A  6  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU540600[8]  A2  A  100  110  55  65  N/A  6  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU543300[8]  A2  A  100  110  55  65  N/A  3  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU543300[8]  A2  A  100  110  55  65  N/A  3  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU543300[8]  A2  A  100  110  55  65  N/A  3  12S  35  I.V. Pole U.V. Light Canopy Connection 
NU543400[8]  A2  A  100  110  55  65  N/A  4  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU543400[8]  A2  A  100  110  55  65  N/A  4  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU543400[8]  A2  A  100  110  55  65  N/A  4  12S  35  I.V. Pole U.V. Light Canopy Connection 
NU543500[8]  A2  A  100  110  57  67  N/A  5  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU543500[8]  A2  A  100  110  57  67  N/A  5  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU543500[8]  A2  A  100  110  57  67  N/A  5  12S  35  I.V. Pole U.V. Light Canopy Connection 
NU543600[8]  A2  A  100  110  55  65  N/A  6  8S  35  U.V. Light I.V. Pole Canopy Connection 
NU543600[8]  A2  A  100  110  55  65  N/A  6  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU543600[8]  A2  A  100  110  55  65  N/A  6  12S  35  I.V. Pole U.V. Light Canopy Connection 
NU545300[9]  A2  A  100  110  55  65  N/A  3  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU545300[9]  A2  A  100  110  55  35  N/A  3  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU545400[10]  A2  A  100  110  55  65  N/A  3  8S  S  I.V. Pole U.V. Light Canopy Connection 
NU545400[10]  A2  A  100  110  55  65  N/A  4  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU545500[11]  A2  A  100  110  55  65  N/A  5  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU545500[11]  A2  A  100  110  55  65  N/A  5  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU545600[12]  A2  A  100  110  55  65  N/A  6  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU545600[12]  A2  A  100  110  55  65  N/A  6  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU560400[13]  B2  A  100  110  55  65  859@1.8  4  8S  36  I.V. Pole U.V. Light 
NU560600[14]  B2  A  100  110  55  35  1373@2.0  6  8S  36  I.V. Pole U.V. Light 
NU565400[13]  B2  A  100  110  55  65  859@1.8  4  8S  36  I.V. Pole U.V. Light 
NU565600[14]  B2  A  100  110  55  65  1373@2.0  6  8S  36  I.V. Pole U.V. Light 
NU677400[15]  A2  A  100  110  55  65  N/A  4  12S  38  Canopy Connection U.V. Light 
NU677500[16]  A2  A  100  110  55  65  N/A  5  12S  38  Canopy Connection U.V. Light 
NU677600[17]  A2  A  100  110  55  65  N/A  6  12S  38  Canopy Connection U.V. Light 
NUFH677500[18]  A2  A  100  110  55  65  N/A  5  12S  38  U.V. Light 
[1] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening for appropriate correction factor, if applicable, without adjusting cabinet airflow balance. [2] Listed model may be suffixed with E. Certified for use with power supplies of 230V/50Hz and 230V/60Hz. [3] Beginning with Series 60 and higher. Downflow nominal setpoint of 70 fpm was established 4 inches above the plane of the window with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492011. This cabinet qualifies with the optional DC motor installed. Listed model may be prefaced with Q, N, L or S (i.e. NUQ427400). [4] Beginning with series 40 and higher. Downflow nominal setpoint of 60 fpm was established 4 inches above the plane of the window with the I.V. Pole and U.V. Light removed. This cabinet qualifies with the optional DC motor installed. Listed model may be prefaced with Q, N, L or S (i.e. NUQ427600). This cabinet model was Certified to NSF/ANSI 492011. [5] Listed model designation NU 435 without U.V. Light option. Listed model may be prefaced with Q, N, L or S (i.e. NUQ435400). [6] Beginning with series 60 and higher. The downflow nominal set point of 60 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model qualifies only with the armrest installed. This cabinet qualifies with the optional DC motor installed. This cabinet model was Certified to NSF/ANSI 492009. Listed model may be prefaced with Q, N, L, or S (i.e. NUQ430400). [7] Beginning with series 60 and higher. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and the U.V. Light removed. This cabinet model qualifies only with the armrest installed. This cabinet qualifies with the optional DC motor installed. This cabinet model was Certified to NSF/ANSI 492009. Listed model may be prefaced with Q, N, L, or S (i.e. NUQ430600). [8] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. Listed model may be prefaced with Q, N, L, or S (e.g. NUQ540300). Certified for use with power supplies of 100V/50Hz, 100V/60Hz, 230V/50Hz, and 230V/60Hz. [9] Beginning with serial number 170844102115. Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492014. Listed model may be prefaced with Q, N, L, or S (e.g. NUQ540300). Certified for use with power supplies of 100V/50Hz, 100V/60Hz, 230V/50Hz, and 230V/60Hz. [10] Beginning with serial number 170847102115. Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492014. Listed model may be prefaced with Q, N, L, or S (e.g. NUQ540300). Certified for use with power supplies of 100V/50Hz, 100V/60Hz, 230V/50Hz, and 230V/60Hz. [11] Beginning with serial number 170845102115. Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492014. Listed model may be prefaced with Q, N, L, or S (e.g. NUQ540300). Certified for use with power supplies of 100V/50Hz, 100V/60Hz, 230V/50Hz, and 230V/60Hz. [12] Beginning with serial number 170846102115. Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492014. Listed model may be prefaced with Q, N, L, or S (e.g. NUQ540300). Certified for use with power supplies of 100V/50Hz, 100V/60Hz, 230V/50Hz, and 230V/60Hz. [13] Beginning with serial number 179773041817. Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492016. Certified for use with power supplies of 115V/60Hz, 230V/50Hz and 230V/60Hz. The concurrent balance value was taken with the NSF exhaust system conforming to ASHRAE. However, differences due to the exhaust system, cabinet and instrumentation may be present and product specification information should be obtained from the manufacturer. [14] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492016. Certified for use with power supplies of 115V/60Hz, 230V/50Hz and 230V/60Hz. The concurrent balance value was taken with the NSF exhaust system conforming to ASHRAE. However, differences due to the exhaust system, cabinet and instrumentation may be present and product specification information should be obtained from the manufacturer. [15] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492011. NuAire canopy model NU911402 was evaluated as part of the NSF Certification testing procedures. [16] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492011. NuAire canopy model NU911502 was evaluated as a part of the NSF Certification testing procedures. [17] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492011. NuAire canopy model NU911602 was evaluated as part of the NSF Certification testing procedures. [18] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. NuAire canopy model NU911502 was evaluated as a part of the NSF Certification testing procedures. NUFH677500 is a modified version of NU677500, including feed hoppers. 
Facility : # 2 Plymouth, MN
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

NU427400[1] [2] [3]  B1  A  100  110  65  75  282 @ 0.7  4  8S  32  BagIn/ BagOut I.V. Pole U.V. Light 
NU427600[1] [2] [4]  B1  A  100  110  65  75  474 @ 0.9  6  8S  32  I.V. Pole U.V. Light BagIn/ BagOut 
NU430400[1] [2] [5] [6]  B2  A  100  110  55  65  785 @ 1.7  4  8S  32  BagIn/ BagOut I.V. Pole U.V. Light 
NU430600[1] [2] [5] [7]  B2  A  100  110  55  65  1250 @ 1.8  6  8S  32  BagIn/ BagOut I.V. Pole U.V. Light 
NU540300[8]  A2  A  100  110  55  65  N/A  3  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU540300[8]  A2  A  100  110  55  65  N/A  3  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU540400[8]  A2  A  100  110  55  65  N/A  4  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU540400[8]  A2  A  100  110  55  65  N/A  4  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU540500[8]  A2  A  100  110  57  67  N/A  5  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU540500[8]  A2  A  100  110  57  67  N/A  5  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU540600[8]  A2  A  100  110  55  65  N/A  6  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU540600[8]  A2  A  100  110  55  65  N/A  6  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU543300[8]  A2  A  100  110  55  65  N/A  3  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU543300[8]  A2  A  100  110  55  65  N/A  3  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU543300[8]  A2  A  100  110  55  65  N/A  3  12S  35  I.V. Pole U.V. Light Canopy Connection 
NU543400[8]  A2  A  100  110  55  65  N/A  4  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU543400[8]  A2  A  100  110  55  65  N/A  4  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU543400[8]  A2  A  100  110  55  65  N/A  4  12S  35  I.V. Pole U.V. Light Canopy Connection 
NU543500[8]  A2  A  100  110  57  67  N/A  5  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU543500[8]  A2  A  100  110  57  67  N/A  5  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU543500[8]  A2  A  100  110  57  67  N/A  5  12S  35  I.V. Pole U.V. Light Canopy Connection 
NU543600[8]  A2  A  100  110  55  65  N/A  6  8S  35  U.V. Light I.V. Pole Canopy Connection 
NU543600[8]  A2  A  100  110  55  65  N/A  6  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU543600[8]  A2  A  100  110  55  65  N/A  6  12S  35  I.V. Pole U.V. Light Canopy Connection 
NU545300[9]  A2  A  100  110  55  65  N/A  3  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU545300[9]  A2  A  100  110  55  35  N/A  3  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU545400[10]  A2  A  100  110  55  65  N/A  3  8S  S  I.V. Pole U.V. Light Canopy Connection 
NU545400[10]  A2  A  100  110  55  65  N/A  4  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU545500[11]  A2  A  100  110  55  65  N/A  5  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU545500[11]  A2  A  100  110  55  65  N/A  5  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU545600[12]  A2  A  100  110  55  65  N/A  6  8S  35  I.V. Pole U.V. Light Canopy Connection 
NU545600[12]  A2  A  100  110  55  65  N/A  6  10S  35  I.V. Pole U.V. Light Canopy Connection 
NU560400[13]  B2  A  100  110  55  65  859@1.8  4  8S  36  I.V. Pole U.V. Light 
NU560600[14]  B2  A  100  110  55  35  1373@2.0  6  8S  36  I.V. Pole U.V. Light 
NU565400[13]  B2  A  100  110  55  65  859@1.8  4  8S  36  I.V. Pole U.V. Light 
NU565600[14]  B2  A  100  110  55  65  1373@2.0  6  8S  36  I.V. Pole U.V. Light 
NU677400[15]  A2  A  100  110  55  65  N/A  4  12S  38  Canopy Connection U.V. Light 
NU677500[16]  A2  A  100  110  55  65  N/A  5  12S  38  Canopy Connection U.V. Light 
NU677600[17]  A2  A  100  110  55  65  N/A  6  12S  38  Canopy Connection U.V. Light 
NUFH677500[18]  A2  A  100  110  55  65  N/A  5  12S  38  U.V. Light 
[1] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening for appropriate correction factor, if applicable, without adjusting cabinet airflow balance. [2] Listed model may be suffixed with E. Certified for use with power supplies of 230V/50Hz and 230V/60Hz. [3] Beginning with Series 60 and higher. Downflow nominal setpoint of 70 fpm was established 4 inches above the plane of the window with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492011. This cabinet qualifies with the optional DC motor installed. Listed model may be prefaced with Q, N, L or S (i.e. NUQ427400). [4] Beginning with series 40 and higher. Downflow nominal setpoint of 60 fpm was established 4 inches above the plane of the window with the I.V. Pole and U.V. Light removed. This cabinet qualifies with the optional DC motor installed. Listed model may be prefaced with Q, N, L or S (i.e. NUQ427600). This cabinet model was Certified to NSF/ANSI 492011. [5] Listed model designation NU 435 without U.V. Light option. Listed model may be prefaced with Q, N, L or S (i.e. NUQ435400). [6] Beginning with series 60 and higher. The downflow nominal set point of 60 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model qualifies only with the armrest installed. This cabinet qualifies with the optional DC motor installed. This cabinet model was Certified to NSF/ANSI 492009. Listed model may be prefaced with Q, N, L, or S (i.e. NUQ430400). [7] Beginning with series 60 and higher. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and the U.V. Light removed. This cabinet model qualifies only with the armrest installed. This cabinet qualifies with the optional DC motor installed. This cabinet model was Certified to NSF/ANSI 492009. Listed model may be prefaced with Q, N, L, or S (i.e. NUQ430600). [8] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. Listed model may be prefaced with Q, N, L, or S (e.g. NUQ540300). Certified for use with power supplies of 100V/50Hz, 100V/60Hz, 230V/50Hz, and 230V/60Hz. [9] Beginning with serial number 170844102115. Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492014. Listed model may be prefaced with Q, N, L, or S (e.g. NUQ540300). Certified for use with power supplies of 100V/50Hz, 100V/60Hz, 230V/50Hz, and 230V/60Hz. [10] Beginning with serial number 170847102115. Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492014. Listed model may be prefaced with Q, N, L, or S (e.g. NUQ540300). Certified for use with power supplies of 100V/50Hz, 100V/60Hz, 230V/50Hz, and 230V/60Hz. [11] Beginning with serial number 170845102115. Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492014. Listed model may be prefaced with Q, N, L, or S (e.g. NUQ540300). Certified for use with power supplies of 100V/50Hz, 100V/60Hz, 230V/50Hz, and 230V/60Hz. [12] Beginning with serial number 170846102115. Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492014. Listed model may be prefaced with Q, N, L, or S (e.g. NUQ540300). Certified for use with power supplies of 100V/50Hz, 100V/60Hz, 230V/50Hz, and 230V/60Hz. [13] Beginning with serial number 179773041817. Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492016. Certified for use with power supplies of 115V/60Hz, 230V/50Hz and 230V/60Hz. The concurrent balance value was taken with the NSF exhaust system conforming to ASHRAE. However, differences due to the exhaust system, cabinet and instrumentation may be present and product specification information should be obtained from the manufacturer. [14] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492016. Certified for use with power supplies of 115V/60Hz, 230V/50Hz and 230V/60Hz. The concurrent balance value was taken with the NSF exhaust system conforming to ASHRAE. However, differences due to the exhaust system, cabinet and instrumentation may be present and product specification information should be obtained from the manufacturer. [15] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492011. NuAire canopy model NU911402 was evaluated as part of the NSF Certification testing procedures. [16] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492011. NuAire canopy model NU911502 was evaluated as a part of the NSF Certification testing procedures. [17] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492011. NuAire canopy model NU911602 was evaluated as part of the NSF Certification testing procedures. [18] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492012. NuAire canopy model NU911502 was evaluated as a part of the NSF Certification testing procedures. NUFH677500 is a modified version of NU677500, including feed hoppers. 
Facility : Shanghai, China
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

NU543300[1]  A2  A  100  110  55  65  N/A  3  10S  35  Canopy Connection I.V. Pole U.V. Light 
NU543400[1]  A2  A  100  110  55  65  N/A  4  10S  35  Canopy Connection I.V. Pole U.V. Light 
NU543500[2]  A2  A  100  110  57  67  N/A  5  10S  35  Canopy Connection I.V. Pole U.V. Light 
NU543600[1]  A2  A  100  110  55  65  N/A  6  10S  35  Canopy Connection I.V. Pole U.V. Light 
[1] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 60 fpm was established 4 inches above the bottom of the sash with the I.V. pole and U.V. light removed. This cabinet model was Certified to NSF 492012. Listed model may be prefaced with Q, N, L, or S (e.g. NUQ540300). Listed model may be suffixed with J, K, or S (e.g. NU543400S). Certified for use with power supplies of 100V/50Hz, 100V/60Hz, 230V/50Hz, and 230V/60Hz. [2] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 62 fpm was established 4 inches above the bottom of the sash with the I.V. pole and U.V. light removed. This cabinet model was Certified to NSF 492012. Listed model may be prefaced with Q, N, L, or S (e.g. NUQ540300). Listed model may be suffixed with J, K, or S (e.g. NU543400S). Certified for use with power supplies of 100V/50Hz, 100V/60Hz, 230V/50Hz, and 230V/60Hz. 
Qingdao Haier Biomedical Co., Ltd.  
No. 280 Fengyuan Road  
HighTech Zone  
Qingdao 266111  
China  
86 176 6092 6003 
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

HR1800IIA2N[1] [2]  A2  A  100  110  65  75  N/A  6  8S  27  U.V. Light Casters Hgt Adjustable Stand 
[1] Beginning with serial number BE0GGBEB200QHM7P0001. Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted access opening (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 70 fpm was established U.V. Light removed. This cabinet is Certified to NSF/ANSI 492019. [2] This cabinet model was Certified for use with power supplies of 100V/60Hz, 115V/60Hz, 230V/50Hz, 230V/60Hz 
Shanghai Lishen Scientific Equipment Co., Ltd.  
No. 6788 SongZe Avenue  
QingPu, Shanghai  
201706  
China  
86 215 986 7888  
Visit this company's website 
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

HFsafe1200LC A2[1]  A2  A  100  110  64  74  N/A  4  8S  27  U.V. Light 
HFsafe1200LC N[1] [2]  A2  A  100  110  64  74  N/A  4  8S  27  U.V. Light 
[1] Beginning with serial number 051312LCJ0003EUS. Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (80 mm high) access opening (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow Certified to NSF/ANSI 492018. Approved metric value for cabinet width is 1250 mm, window height is 200 mm, and bench height max is 690 mm. [2] Approved for alternate power mode of 230V/50Hz. 
Suzhou Antai Airtech Co., Ltd.  
No. 2 Weixin Road, SIP  
Suzhou, Jiangsu 215122  
China  
86 512 6552 5002  
Visit this company's website 
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

BSC1301 II A2[1]  A2  A  100  110  64  74  N/A  4  10S  24  U.V. Light Canopy Connection 
BSC1302 II A2[1] [2]  A2  A  100  110  64  74  N/A  4  10S  24  U.V. Light Canopy Connection 
[1] Beginning with serial number J14121740. Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using manufacturer's recommended alternate method with thermal anemometer in a constricted (80 mm high) access opening and applying appropriate correction factor without adjusting cabinet airflow balance. The downflow nominal setpoint of 69 fpm was established 4 inches above the bottom of the sash with the U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492014. Approved metric value for cabinet width is 1270 mm, window height is 250 mm, and bench height max is 610 mm. [2] Certified for use with power supplies of 230V/50Hz and 230V/60Hz. 
Thermo Fisher Scientific  
RobertBoschStr. 1  
D63505 Langenselbold  
Germany  
49 61 813 5277  
Visit this company's website 
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

1305[1] [2]  A2  A  100  110  58  68  N/A  4  8S  36  Armrest I.V. Pole U.V. Light 
1306[1] [2] [3]  A2  A  100  110  58  58  N/A  4  8S  36  Armrest I.V. Pole U.V. Light 
1307[1] [4]  A2  A  100  110  58  68  N/A  6  8S  36  Armrest I.V. Pole U.V. Light 
1308[1] [3] [4]  A2  A  100  110  58  68  N/A  6  8S  36  Armrest I.V. Pole U.V. Light 
1320[5] [6] [7]  A2  B  100  110  58  68  N/A  3  10S  36  Armrest I.V. Pole U.V. Light 
1321[5] [6] [7]  A2  A  100  110  58  68  N/A  3  10S  36  Armrest I.V. Pole U.V. Light 
1322[5] [6] [7]  A2  B  100  110  58  68  N/A  3  10S  36  Armrest I.V. Pole U.V. Light 
1323[5] [6] [7]  A2  A  100  110  58  68  N/A  3  10S  36  Armrest U.V. Light I.V. Pole 
1325[1] [2]  A2  A  100  110  58  68  N/A  4  8S  36  Armrest I.V. Pole U.V. Light 
1326[1] [2] [3]  A2  A  100  110  58  68  N/A  4  8S  36  Armrest I.V. Pole U.V. Light 
1327[1] [4]  A2  A  100  110  58  68  N/A  6  8S  36  Armrest I.V. Pole U.V. Light 
1328[1] [3] [4]  A2  A  100  110  58  68  N/A  6  8S  36  Armrest I.V. Pole U.V. Light 
1330[3] [5] [6] [7]  A2  B  100  110  58  68  N/A  3  10S  36  Armrest I.V. Pole U.V. Light 
1331[3] [5] [6] [7]  A2  A  100  110  58  68  N/A  3  10S  36  Armrest I.V. Pole U.V. Light 
1332[3] [5] [6] [7]  A2  B  100  110  58  68  N/A  3  10S  36  Armrest I.V. Pole U.V. Light 
1333[3] [5] [6] [7]  A2  A  100  110  58  68  N/A  3  10S  36  Armrest I.V. Pole U.V. Light 
1335[1] [2]  A2  A  100  110  58  68  N/A  4  8S  36  Armrest I.V. Pole U.V. Light 
1336[1] [2] [3]  A2  A  100  110  58  68  N/A  4  8S  36  Armrest I.V. Pole U.V. Light 
1337[1] [4]  A2  A  100  110  58  68  N/A  6  8S  36  Armrest I.V. Pole U.V. Light 
1338[1] [3] [4]  A2  A  100  110  58  68  N/A  6  8S  36  Armrest I.V. Pole U.V. Light 
1339[5] [6] [7]  A2  B  100  110  58  68  N/A  3  8S  36  Armrest I.V. Pole U.V. Light 
1340[5] [6] [7]  A2  A  100  110  58  68  N/A  3  8S  36  Armrest I.V. Pole U.V. Light 
1341[5] [6] [7]  A2  B  100  110  58  68  N/A  3  8S  36  Armrest I.V. Pole U.V. Light 
1342[5] [6] [7]  A2  A  100  110  58  68  N/A  3  8S  36  Armrest I.V. Pole U.V. Light 
1345[2] [8]  A2  A  100  110  58  68  N/A  4  10S  36  Armrest I.V. Pole U.V. Light 
1346[2] [8] [9]  A2  A  100  110  58  68  N/A  4  10S  36  Armrest I.V. Pole U.V. Light 
1347[4] [8]  A2  A  100  110  58  68  N/A  6  10S  36  Armrest I.V. Pole U.V. Light 
1348[4] [8] [9]  A2  A  100  110  58  68  N/A  6  10S  36  Armrest I.V. Pole U.V. Light 
1350[5] [6] [10]  A2  B  100  110  58  68  N/A  5  8S  36  Armrest I.V. Pole U.V. Light 
1351[5] [6] [10]  A2  A  100  110  58  68  N/A  5  8S  36  Armrest I.V. Pole U.V. Light 
1352[5] [6] [10]  A2  B  100  110  58  68  N/A  5  8S  36  Armrest I.V. Pole U.V. Light 
1353[5] [6] [10]  A2  A  100  110  58  68  N/A  5  8S  36  Armrest I.V. Pole U.V. Light 
1354[2] [11]  A2  A  100  110  58  68  N/A  4  10S  36  Armrest I.V. Pole U.V. Light 
1355[2] [8]  A2  A  100  110  58  68  N/A  4  10S  36  Armrest I.V. Pole U.V. Light 
1356[2] [8] [9]  A2  A  100  110  58  68  N/A  4  10S  36  Armrest I.V. Pole U.V. Light 
1357[4] [8]  A2  A  100  110  58  68  N/A  6  10S  36  Armrest I.V. Pole U.V. Light 
1358[4] [8] [9]  A2  A  100  110  58  68  N/A  6  10S  36  Armrest I.V. Pole U.V. Light 
1359[4] [8]  A2  A  100  110  58  68  N/A  6  10S  36  Armrest I.V. Pole U.V. Light 
1366[3] [5] [6] [10]  A2  B  100  110  58  68  N/A  5  8S  36  Armrest I.V. Pole U.V. Light 
1367[3] [5] [6] [10]  A2  A  100  110  58  68  N/A  5  8S  36  Armrest U.V. Light I.V. Pole 
1368[5] [6] [10]  A2  B  100  110  58  68  N/A  5  10S  36  Armrest I.V. Pole U.V. Light 
1369[5] [6] [10]  A2  A  100  110  58  68  N/A  5  10S  36  Armrest U.V. Light I.V. Pole 
1370[5] [6] [10]  A2  B  100  110  58  68  N/A  5  10S  36  Armrest I.V. Pole U.V. Light 
1371[5] [6] [10]  A2  A  100  110  58  68  N/A  5  10S  36  Armrest I.V. Pole U.V. Light 
1372[3] [5] [6] [10]  A2  B  100  110  58  68  N/A  5  8S  36  Armrest U.V. Light I.V. Pole 
1373[3] [5] [6] [10]  A2  A  100  110  58  68  N/A  5  8S  36  Armrest I.V. Pole U.V. Light 
1375[2] [8]  A2  A  100  110  58  68  N/A  4  10S  36  Armrest I.V. Pole U.V. Light 
1376[2] [8] [9]  A2  A  100  110  58  68  N/A  4  10S  36  Armrest I.V. Pole U.V. Light 
1377[4] [8]  A2  A  100  110  58  68  N/A  6  10S  36  Armrest I.V. Pole U.V. Light 
1378[4] [8] [9]  A2  A  100  110  58  68  N/A  6  10S  36  Armrest I.V. Pole U.V. Light 
1380[3] [5] [6] [10]  A2  B  100  110  58  68  N/A  5  10S  36  Armrest I.V. Pole U.V. Light 
1381[3] [5] [6] [10]  A2  A  100  110  58  68  N/A  5  10S  36  Armrest I.V. Pole U.V. Light 
1382[3] [5] [6] [10]  A2  B  100  110  58  68  N/A  5  10S  36  Armrest I.V. Pole U.V. Light 
1383[3] [5] [6] [10]  A2  A  100  110  58  68  N/A  5  10S  36  Armrest I.V. Pole U.V. Light 
1384[2] [11]  A2  A  100  110  58  68  N/A  4  10S  36  Armrest I.V. Pole U.V. Light 
1385[2] [8]  A2  A  100  110  58  68  N/A  4  10S  36  Armrest I.V. Pole U.V. Light 
1386[2] [8] [9]  A2  A  100  110  58  68  N/A  4  10S  36  Armrest I.V. Pole U.V. Light 
1387[4] [8]  A2  A  100  110  58  68  N/A  6  10S  36  Armrest I.V. Pole U.V. Light 
1388[4] [8] [9]  A2  A  100  110  58  68  N/A  6  10S  36  Armrest I.V. Pole U.V. Light 
1389[4] [11]  A2  A  100  110  58  68  N/A  6  10S  36  Armrest I.V. Pole U.V. Light 
1390[3] [5] [6] [7]  A2  B  100  110  58  68  N/A  3  8S  36  Armrest I.V. Pole U.V. Light 
1391[3] [5] [6] [7]  A2  A  100  110  58  68  N/A  3  8S  36  Armrest I.V. Pole U.V. Light 
1392[3] [5] [6] [7]  A2  B  100  110  58  68  N/A  3  8S  36  Armrest I.V. Pole U.V. Light 
1393[3] [5] [6] [7]  A2  A  100  110  58  68  N/A  3  8S  36  Armrest I.V. Pole U.V. Light 
1395[2] [9] [12]  A2  A  100  110  58  68  N/A  4  8S  36  Armrest I.V. Pole U.V. Light 
1396[1] [2] [3]  A2  A  100  110  58  68  N/A  4  8S  36  Armrest I.V. Pole U.V. Light 
1397[4] [9] [12]  A2  A  100  110  58  68  N/A  6  8S  36  Armrest I.V. Pole U.V. Light 
1398[1] [3] [4]  A2  A  100  110  58  68  N/A  6  8S  36  Armrest I.V. Pole U.V. Light 
[1] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal setpoint was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal setpoint of 63 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492009. [2] Beginning April 20, 2020 a new fixed height base stand GT221201 and adjustable height base stand GT221189 along with 4ft cross brace GT221195 were approved for use with all 1300 series models manufactured in Marietta OH. Casters GT221199 is an approved option for both stands. [3] Certified for use with power supply of 230V/50Hz and 230V/60Hz. [4] Beginning April 20, 2020 a new fixed height base stand GT221201 and adjustable height base stand GT221189 along with 6ft cross brace GT221196 were approved for use with all 1300 series models manufactured in Marietta OH. Casters GT221199 is an approved option for both stands. [5] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's alternate recomended method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer's operator's manual for appropriate correction factor), without adjusting cabinet airflow balance. Downflow nominal set point was established with I.V.Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492009. [6] May be used with alternate, adjustable height electric stand (maximum bench height of 35 inches). [7] Beginning April 20, 2020 a new fixed height base stand GT221201 and adjustable height base stand GT221189 along with 3ft cross brace GT221193 were approved for use with all 1300 series models manufactured in Marietta OH. Casters GT221199 is an approved option for both stands. [8] Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 63 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492009. Certified for use with power supply of 230V/50Hz and 230V/60Hz. [9] Certified for use with power supply of 220 V / 50 Hz and 220 V / 60 Hz. [10] Beginning April 20, 2020 a new fixed height base stand GT221201 and adjustable height base stand GT221189 along with 5ft cross brace GT221197 were approved for use with all 1300 series models manufactured in Marietta OH. Casters GT221199 is an approved option for both stands. [11] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal setpoint was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal setpoint of 63 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492009. Certified for use with power supply of 220V/50Hz. [12] Beginning with serial number 1881221 and 1881081 for models 1395 and 1397, respectively. Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 63 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492009. 
Facility : Suzhou, China
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

1304[1] [2]  A2  A  100  110  58  68  N/A  4  8S  36  Armrest I.V. Pole U.V. Light 
1309[1] [2]  A2  A  100  110  58  68  N/A  6  8S  S  Armrest I.V. Pole U.V. Light 
1374[1] [2] [3]  A2  A  100  110  58  68  N/A  4  10S  36  Armrest I.V. Pole U.V. Light 
1379[1] [2] [4]  A2  A  100  110  58  68  N/A  6  10S  36  Armrest I.V. Pole U.V. Light 
1384[1] [2]  A2  A  100  110  58  68  N/A  4  10S  36  Armrest I.V. Pole U.V. Light 
1389[1] [2]  A2  A  100  110  58  68  N/A  6  10S  36  Armrest I.V. Pole U.V. Light 
1394[1] [2]  A2  A  100  110  58  68  N/A  4  8S  36  Armrest I.V. Pole U.V. Light 
1399[1] [2]  A2  A  100  110  58  68  N/A  6  8S  36  Armrest I.V. Pole U.V. Light 
[1] Models can be provided with Type F power cord and internal receptacles and identified as 1304F, 1309F, 1374F, 1379F, 1384F, 1389F, 1394F, or 1399F; with Type G power cord and internal receptacles and identified as 1304G, 1309G, 1374G, 1379G, 1384G, 1389 G, 1394G, or 1399G; with Type M power cord and internal receptacles and identified as 1304M, 1309M, 1374M, 1379M, 1384M, 1389M, 1394M, or 1399M; or with the standard Type I power cord and internal receptacles. [2] Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal setpoint was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal setpoint of 63 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492014. Certified for use with power supplies of 220V/50Hz and 220V/60Hz. [3] Beginning with serial number 1374318010704. [4] Beginning with serial number 1379318010304. 
Facility : Langenselbold, Germany
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

HERAsafe 2030i 1.2[1]  A2  A  100  110  58  68  N/A  4  10S  36  Armrest I.V. Pole U.V. Light 
HERAsafe 2030i 1.8[1]  A2  A  100  110  58  68  N/A  6  10S  36  Armrest I.V. Pole U.V. Light 
KS9[2]  A2  A  100  110  65  75  N/A  3  10 10H S 
35  Full Side Windows I.V. Pole U.V. Light Armrest Casters 
KS12[3]  A2  A  100  110  70  80  N/A  4  10 10H S 
34  Full Side Windows U.V. Light I.V. Pole Armrest Casters 
KS15[4]  A2  A  100  110  65  75  N/A  5  10 10H S 
35  Full Side Windows I.V. Pole U.V. Light Armrest Casters 
KS18[5]  A2  A  100  110  65  75  N/A  6  10 10H S 
34  Full Side Windows I.V. Pole U.V. Light Casters 
[1] Beginning with serial number 42294669 for the HERAsafe 2030i 1.2 and serial number 42209172 for the HERAsafe 2030i 1.8. Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer's operator's manual for appropriate correction factor) without adjusting cabinet airflow balance. Downflow nominal set point of 63 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified with both dished and flat work surfaces and for additional inlet power supplies of 230V/50Hz and 230V/60Hz. This cabinet model was Certified to NSF/ANSI 492018. HERAsafe 2030i 1.2 is approved for use with either padded armrests or armrests without padding. HERAsafe 2030i 1.8 is approved for use without armrests, with padded armrests, or with armrests without padding. [2] Beginning with serial number 40329642. Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. This nominal setpoint was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer's operator's manual for appropriate correction factor) without adjusting cabinet airflow balance. This cabinet model was Certified with a 3 section work surface. This cabinet model was Certified to NSF/ANSI 492009. [3] Beginning with serial number 40292176. Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer's operator's manual for appropriate correction factor, if applicable) without adjusting cabinet airflow balance. This cabinet model was Certifed with a 4 section work surface. This cabinet model was Certified to NSF/ANSI 492007. [4] Beginning with serial number 40349384. Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. This nominal setpoint was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer's operator's manual for appropriate correction factor) without adjusting cabinet airflow balance. This cabinet model was Certified with a 5 section work surface. This cabinet model was Certified to NSF/ANSI 492009. [5] Beginning with serial number 40292176. Inflow nominal setpoint of 105 fpm was established with a direct airflow reading instrument. A corresponding inflow nominal setpoint of 105 fpm (corrected to local air density) was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted (3 inch high) access opening (consult manufacturer's operator's manual for appropriate correction factor, if applicable) without adjusting cabinet airflow balance. This cabinet model was Certified with a 6 section work surface. This cabinet model was Certified to NSF/ANSI 492007. 
Thermo Fisher Scientific  
401 Millcreek Road  
Marietta, OH 45750  
United States  
7403734763 
Model Number 
Cabinet Type/Style 
Inflow Velocity (fpm) 
Downflow Velocity (fpm) 
CBV (cfm) at Static Pressure (in w.g.) 
Cabinet Width ft. 
Window Ht/Type in. 
Bench Ht Max in. 
Acceptable Options 

151x[1] [2] [3]  B2  A  100  110  50  60  852 @ 1.8  4  8S  36  BagIn/ BagOut I.V. Pole U.V. Light 
156x[1] [2] [4]  B2  A  100  110  50  60  1265 @ 2.2  6  8S  36  BagIn/ BagOut I.V. Pole U.V. Light 
[1] Beginning with serial number 190500000. Inflow nominal set point of 105 fpm was established with a direct airflow reading instrument. This nominal set point was confirmed using the manufacturer's recommended alternate method with thermal anemometer in a constricted access opening, using the manufacturersupplied probe holder and sash stop (consult manufacturer for appropriate correction factor, if applicable), without adjusting cabinet airflow balance. Downflow nominal set point of 55 fpm was established with I.V. Pole and U.V. Light removed. This cabinet model was Certified to NSF/ANSI 492016. This cabinet model is Certified for use with optional cord & cable portal. [2] Certified for use with power supplies of 100V/50 Hz, 100V/60 Hz, 230V/50 Hz, and 230V/60 Hz. [3] Model 1511 is equivalent to Model 1510, but comes equipped with U.V. lighting (Listed as an acceptable option). Model 1515 is equivalent to model 1510, but is rated for use with power supplies of 230V and 50/60 Hz and is equipped with a Chinesecompatible plug. Model 1516 is equivalent, but is rated for use with power supplies of 230V and 50/60 Hz and is equipped with U.V. lighting and a Chinesecompatible plug. Model 1517 is equivalent, but is rated for use with power supplies of 230V and 50/60 Hz and is equipped with a North American 230 Voltcompatible plug. Model 1517F is equivalent, but is rated for use with power supplies of 230V and 50/60 Hz and is equipped with a Type F electrical plug (Schuko). Model 1517G is equivalent, but is rated for use with power supplies of 230V and 50/60 Hz and is equipped with a Type G electrical plug (UKcompatible). Model 1517M is equivalent, but is rated for use with power supplies of 230V and 50/60 Hz and is equipped with a Type M electrical plug (Indiacompatible). Model 1518 is equivalent, but is rated for use with power supplies of 230V and 50/60 Hz and is equipped with U.V. lighting and a North American 230 Voltcompatible plug.Model 1518F is equivalent, but is rated for use with power supplies of 230V and 50/60 Hz and is equipped with a Type F electrical plug (Schuko). Model 1518G is equivalent, but is rated for use with power supplies of 230V and 50/60 Hz and is equipped with a Type G electrical plug (UKcompatible). Model 1518M is equivalent, but is rated for use with power supplies of 230V and 50/60 Hz and is equipped with a Type M electrical plug (Indiacompatible). [4] Model 1561 is equivalent to Model 1560, but comes equipped with U.V. lighting (Listed as an acceptable option). Model 1562 is equivalent to model 1560, but is rated for use with power supplies of 230V and 50/60 Hz and is equipped with a Chinesecompatible plug. Model 1563 is equivalent, but is rated for use with power supplies of 230V and 50/60 Hz and is equipped with U.V. lighting and a Chinesecompatible plug. Model 1564 is equivalent, but is rated for use with power supplies of 230V and 50/60 Hz and is equipped with a North American 230 Voltcompatible plug. Model 1564F is equivalent, but is rated for use with power supplies of 230V and 50/60 Hz and is equipped with a Type F electrical plug (Schuko). Model 1564G is equivalent, but is rated for use with power supplies of 230V and 50/60 Hz and is equipped with a Type G electrical plug (UKcompatible). Model 1564M is equivalent, but is rated for use with power supplies of 230V and 50/60 Hz and is equipped with a Type M electrical plug (Indiacompatible). Model 1565 is equivalent, but is rated for use with power supplies of 230 V and 50/60 Hz and is equipped with U.V. lighting and a North American 230 Voltcompatible plug. Model 1565F is equivalent, but is rated for use with power supplies of 230V and 50/60 Hz and is equipped with a Type F electrical plug (Schuko). Model 1565G is equivalent, but is rated for use with power supplies of 230V and 50/60 Hz and is equipped with a Type G electrical plug (UKcompatible). Model 1565M is equivalent, but is rated for use with power supplies of 230V and 50/60 Hz and is equipped with a Type M electrical plug (Indiacompatible). 
Number of matching Manufacturers is 18
Number of matching Products is 308
Processing time was 0 seconds